Methods and apparatus for electro-meridian diagnostics

ABSTRACT

A current sensor may take measurements of electrical currents that flow between two limbs of a mammal through at least a portion of the mammal’s torso. The current measurements may be taken during a single diagnostic session while a ground electrode contacts the body at a preferably distal location on a limb and a probe electrode is sequentially placed at different locations on distal portions of other limbs. Each of the current delivery locations may be an acupuncture point. An electrical current state for the diagnostic session may be calculated. This state may consist of current ranges for one or more electrical currents that are measured during the session. A lookup table may be employed to determine one or more medical conditions that are indicated by the current state. Alternatively, a trained machine learning model may predict, based on the measured currents, one or more medical conditions.

RELATED APPLICATIONS

This application claims the benefit of co-pending U.S Pat. Application Serial No. 16/836,806, filed Mar. 31, 2020, which claims the benefit of U.S Provisional Pat. Application Serial No. 62/828,206, filed Apr. 2, 2019, both of which are incorporated herein by reference in their entirety.

SUMMARY OF THE INVENTION

In illustrative implementations, a diagnostic system employs a current sensor to screen for and to detect a wide variety of medical conditions. The current sensor may take measurements of small applied electrical currents that flow between a probe electrode and a ground electrode, while an animal (e.g. human or mammal) holds the ground electrode in one hand (or the ground electrode electrically contacts the animal body at a distal portion of a limb) and the probe electrode is sequentially placed at different locations on the patient’s two feet and on the patient’s other forearm. These cross-body currents may flow through at least a portion of the patient’s torso. The ground electrode may be switched from one hand to another, to enable current measurements to be taken for both forearms.

The measurements of electrical current may be taken during a single diagnostic session. Each of the measurement locations may be an acupuncture point.

Based on the measurements, an electrical current state for the diagnostic session may be calculated. This state may consist of: (a) a current range for an electrical current that is measured during the session; or (b) current ranges for respective currents that are measured during the session. A lookup table may be employed to determine one or more medical conditions that are indicated by the current state. Alternatively, a trained machine learning model may predict, based on the measured currents, one or more medical conditions.

In some cases, the diagnostic system determines whether or not a patient has a viral infection and whether or not a patient has a bacterial infection, based on electrical current measurements that take only a few minutes. This ability to quickly and accurately detect and differentiate between viral and bacterial infections enables the diagnostic system to be used as a mass-scale, rapid screening tool in a viral or bacterial epidemic. For instance, during the COVID-19 pandemic, the diagnostic tool may be used to quickly determine whether a patient has a viral or bacterial infection or both, and if a viral infection is indicated, to refer the patient for a panel of respiratory virus tests, including a COVID-19 assay.

In some cases, the ground electrode and probe electrode are attached to flexible wires and are free to move relative to each other.

Alternatively, in some cases, the ground electrode and probe electrode are rigid parts of a single rigid structure and thus are in a fixed position relative to each other. The rigid structure may be configured to also serve as a case for a smartphone. The rigid structure may enable a patient to hold both the probe electrode and ground electrode in one hand. For instance, the patient may hold the rigid structure in such a way that the ground electrode of the rigid structure is pressed against the palm of one hand, while the patient sequentially presses the probe electrode at different points on the patient’s left foot, right foot and other forearm.

In some cases, one or more pressure sensors measure how much pressure is being applied to the probe and/or ground electrodes. These pressure readings, as well as current readings by the current sensor, may be employed to determine whether the electrodes are being pressed properly against the patient’s skin to achieve sufficient conductance for accurate measurements. In some cases, the ground electrode and probe electrode are rigid parts of a single rigid structure and thus are in a fixed position relative to each other, except for any movement that is due solely to displacements that occur within one or more pressure sensors.

A user interface (UI) may present to a user: (a) information about the measurements; (b) a diagnosis or tentative diagnosis; and/or (c) a recommendation for further medical testing. In addition, the UI may provide real-time audiovisual feedback to a user regarding whether the electrodes are being used properly.

The Summary and Abstract sections and the title of this document: (a) do not limit this invention; (b) are intended only to give a general introduction to some illustrative implementations of this invention; (c) do not describe all of the details of this invention; and (d) merely describe non-limiting examples of this invention. This invention may be implemented in many other ways. Likewise, the Field of Technology section is not limiting; instead it identifies, in a general, non-exclusive manner, a field of technology to which some implementations of this invention generally relate.

BRIEF DESCRIPTION OF THE DRAWINGS

FIG. 1 shows a current sensor which has ground and probe electrodes that are free to move relative to each other.

FIG. 2 shows electrodes of a current sensor being employed to measure cross-body electrical currents.

FIGS. 3, 4 and 5 show rigid structures that each include both ground and probe electrodes.

FIG. 6 shows a spring-loaded electrode.

FIGS. 7, 8, 9, 10A and 10B show measurement points.

FIG. 11 illustrates cross-body currents.

FIG. 12 is a flowchart for a diagnostic method.

FIG. 13 is a diagram that illustrates a relational database.

FIGS. 14A-D shows measurement points located on representative non-human animals.

The above Figures are not necessarily drawn to scale. The above Figures show illustrative implementations of this invention, or provide information that relates to those implementations. The examples shown in the above Figures do not limit this invention. This invention may be implemented in many other ways.

DESCRIPTION OF THE PREFERRED EMBODIMENT

Although the disclosure hereof is detailed and exact to enable those skilled in the art to practice the invention, the physical embodiments herein disclosed merely exemplify the invention which may be embodied in other specific structures. While the preferred embodiment has been described, the details may be changed without departing from the invention.

Current Sensor

In illustrative implementations of this invention, a current sensor measures what we sometimes call “cross-body” electrical currents. In some cases, the cross-body currents are electrical currents that flow between distal regions of two limbs of a patient, passing through at least a portion of the patient’s torso. In some use scenarios, the current sensor measures a cross-body electrical current that flows between: (a) skin on a patient’s hand; and (b) skin on a foot or ankle of the patient. In other use scenarios, the current sensor measures a cross-body electrical current that flows between: (a) skin of a hand of a patient’s forearm; and (b) skin of a hand or wrist of the patient’s other forearm. In each of the preceding examples, the cross-body electrical current may flow through at least a portion of the patient’s torso. In some use scenarios, the current sensor measures cross- body electrical currents that pass through the sagittal plane and/or transpyloric plane of the patient’s body.

In some use scenarios, the cross-body electrical currents are very small in magnitude. For instance, in some cases, these electrical currents are in a range from 0.1 microamperes to 500 microamperes, or in a range from 0.1 microamperes to 300 microamperes. In some implementations, the current sensor measures cross-body currents while: (a) a patient holds a ground electrode; and (b) a probe electrode is positioned at different locations on the patient’s skin to apply a diagnostic electric current (preferably about 200 to about 500 microamperes) to such locations for a predetermined amount of time, such as about three to about five seconds. For instance, cross-body currents may be measured while the probe electrode is located at 24 different locations on the patient’s limbs, one location at a time. The 24 probe locations may consist of: (a) six locations on the right foot and six corresponding locations on the left foot; and (b) six locations on the right hand (or right wrist) and six corresponding locations on the left hand (or left wrist) .

In some cases, the current sensor has ground and probe electrodes that are not in a fixed position relative to each other. Put differently, in some cases, the ground and probe electrodes are free to move relative to each other.

FIG. 1 shows a current sensor which has ground and probe electrodes that are not in a fixed position relative to each other. In FIG. 1 , a current sensing system includes a ground electrode 103, a probe electrode 101, and module 104. The ground and probe electrodes are each connected to flexible wires and may move relative to each other. Ground electrode 103 is configured to be held by a patient directly against the skin of the patient’s hand, while the current sensor measures cross-body currents that flow through the patient. Probe electrode 101 has a conductive tip 102 that is configured to be pressed directly against the patient’s skin at each of multiple measurement points, one measurement location at a time. The main body of probe electrode 101 (other than conductive tip 101) may be covered by a thin insulative sheath.

In FIG. 1 , wires may electrically connect the ground and probe electrodes with module 104. Module 104 may house (among other things) power circuitry 123, ammeter 122 and a microprocessor 121. The power circuitry 123 may include a power source, a (non-ideal) current source or a (non-ideal) voltage source or may otherwise generate or modulate a cross-body electrical current. Power circuitry 123 may in turn receive power from computer 105.

The cross-body electrical currents (which are generated by the power circuitry 123 and that flow between the ground and probe electrodes through a patient’s body) may be either DC currents (direct currents) or AC currents (alternating currents). In some cases, microprocessor 121 includes a signal generator. This signal generator: (a) may comprise an oscillator, function generator, waveform generator, or digital pattern generator; and (b) may be employed to control timing and duration of a DC or AC cross- body current.

In FIG. 1 , ammeter 122 may comprise any type of current sensor or ammeter, including any type of digital ammeter. For instance, ammeter 122 may employ a shunt resistor to produce an analog voltage that is proportional to current and this voltage may in turn be measured by a digital voltmeter, which employs an ADC (analog- to-digital converter) to convert analog voltage to digital data. In some cases, ammeter 122 includes a current sense amplifier, which comprises a differential amplifier with a matched resistive gain network that monitors current flow by measuring current drop across a sense element, such as a shunt resistor. The current sense amplifier may include an integrated current-sense resistor. In some other cases, ammeter 122 comprises a Hall effect current sensor, transformer current sensor, current clamp sensor, fluxgate transformer current sensor, moving coil ammeter, moving magnet ammeter, or electrodynamic ammeter. Ammeter 122 may produce an analog voltage that is calibrated to be proportional to current, and an ADC may convert this analog voltage to digital data.

In FIG. 1 , ammeter 122 may output digital data that represents measurements of cross-body electrical currents that are taken at different points on the patient’s limbs. Microprocessor 121 may analyze this digital data.

In FIG. 1 , computer 105 controls and interfaces with microprocessor 122, and may further analyze data. Computer 105 may store data in, and access data from, a memory device 124. Computer 105 may interface with a set of input/output (I/O) devices, including a microphone 131, speaker 132, electronic display screen 133 (e.g., a touch screen, computer monitor, or laptop screen), keyboard 134 and mouse 135.

In some use scenarios, a health-care worker holds probe electrode 101 and presses it against different points in the patient’s skin, while the patient holds ground electrode 103. At each of the measurement locations, a cross-body electrical current may be measured. For instance, while the patient holds the ground electrode 103 in the palm of one hand with fingers curling around the ground electrode, the health-care worker may hold probe electrode 101 and press it against a sequence of 24 locations on the patient’s body, one location at a time. As a non-limiting example, the health-care worker may first press probe electrode 101 against six locations on the patient’s right foot, then against six locations on the patient’s left foot, then against six locations on the user’s right hand or wrist, and then against six locations on the user’s left hand or wrist. The current sensor may measure cross-body currents that flow when the probe electrode is at each of these different measurement locations.

In the example shown in FIG. 2 , a health care worker may administer the diagnostic test to a patient. Specifically, in FIG. 2 : (a) a patient may hold ground electrode 103 in the palm of a hand 112 with fingers gripping and wrapped around the ground electrode; while (b) a health care worker (not shown) presses the conductive tip 102 of probe electrode 101 against the skin of the patient’s other forearm 111.

In some use scenarios, a patient may self-administer at least a portion of the diagnostic test. For instance, while the patient holds the ground electrode 103 in one hand, the patient may hold probe electrode 101 in the other hand and may press it against six locations in the patient’s right foot and six locations in the patient’s left foot.

However, the apparatus shown in FIG. 1 is not well-suited for a patient himself or herself to take measurements of cross-body currents that occur at hand or wrist measurement points. This is because it can be difficult for the patient to hold the ground electrode and probe electrode in the same hand while pressing the probe electrode against the hand or wrist of the patient’s other forearm. When the patient grips the ground electrode in the palm of a hand (with fingers wrapped around the ground electrode), it may be difficult for the patient to also hold the probe electrode in the fingers of the same hand.

In some implementations of this invention, this problem is solved by employing a current sensor in which the ground electrode and probe electrode are parts of a single rigid structure and thus are in a fixed position relative to each other. The patient may hold the rigid structure in one hand, with the ground electrode portion of the rigid structure pressed against the palm of that hand, while pressing the probe electrode portion of the rigid structure against the skin of another extremity. For instance, a user may hold the rigid structure in the right hand with the ground electrode pressed against the skin of the right hand, while pressing the probe electrode first against six locations on the right foot, then against six locations on the left foot, and then against six locations on the left hand. Then the user may hold the rigid structure in the left hand, while pressing the probe electrode portion of the rigid structure against six locations on the right hand. At each of the different measurement locations, the current sensor may measure a cross- body electrical current.

FIGS. 3, 4 and 5 show rigid structures that each: (a) are part of a current sensor; and (b) include both ground and probe electrodes.

In the example shown in FIG. 3 , rigid structure 300 is configured to fit tightly around, and to hold in place, a smartphone. Put differently, rigid structure 300 may function in part as a rigid case that partially surrounds, and holds in place, a smartphone. Rigid structure 300 has a back 330 and walls 301. A smartphone may be inserted into a recessed region 340 of structure 300, in such a way that: (a) the smartphone presses against back 330 of structure 300; and (b) lateral movement of the smartphone is constrained by walls 301. Walls 301 may snap-fit around or press tightly against the smartphone, causing the smartphone to remain in recessed region 340 unless a user pulls on the smartphone to remove it from the recessed region. Back 330 has holes 332, 333, in order to reduce the weight of structure 300.

In FIG. 3 , rigid structure 300 also includes a probe electrode 320 and a ground electrode 390. In FIG. 3 , probe electrode 320 and a ground electrode 390 are rigid parts of a single rigid structure and thus are in a fixed position relative to each other. Probe electrode 320 includes a conductive tip 323. Ground electrode 390 (hidden from view in FIG. 3 ) and recessed region 340 are on opposite sides of back 330.

A patient: (a) may hold rigid structure 300 in one hand, in such a way that ground electrode is pressed against skin of the palm of that hand, and (b) may press the conductive tip 323 of probe electrode 322 against locations on the skin of other extremities. For instance, the patient may hold rigid structure 300 in the patient’s left hand, while pressing tip 323 against a sequence of locations, such as six locations on the patient’s right foot, then six locations on the patient’s left foot, and then six locations on the patient’s right forearm. The patient may then hold the rigid structure in the right hand, and press tip 323 against a sequence of six locations on the patient’s left forearm. The current sensor may measure cross-body currents when the probe electrode is at each of these different locations.

FIGS. 4 and 5 show a front view and back view, respectively, of a rigid structure 400. Rigid structure 400 functions in part as a case for a smartphone. Walls 401 and back 490 form a recessed region into which a smartphone 450 may be inserted. Smartphone 450 may include touch screen 451.

In the example shown in FIGS. 4 and 5 , rigid structure 400 includes a probe electrode 420 with a conductive tip 423, and also includes a ground electrode. This ground electrode has six conductive pads 452, 453, 454, 480, 481, 470. Again, a patient may hold rigid structure 400 in one hand, with ground electrode pressed against the skin of the palm of that hand, while pressing the probe electrode 420 against the skin at different locations on other extremities of the patient’s body. The current sensor may measure cross-body currents at each of these measurement locations. An electronics module 460 may include an ADC, other signal processing circuity and a microcontroller. Electronics module 460 may include an ammeter. The hardware and functionality of the ammeter in electronics module 460 may be the same as described above with respect to ammeter 122. An interface module 461 may include electronic components and other circuitry for interfacing with the smartphone. In some cases, interface module 461 is self-cleaning or self-polishing. For instance, interface module 461 may include pliant layers that tend to scrape debris off of conducting electrodes when the smartphone (or other mobile computing device) is being inserted into the recessed region of the rigid structure 400. These pliant layers may comprise Teflon®.

In some use scenarios, it is desirable to measure how forcefully the ground electrode and/or probe electrode are being pressed against skin of the patient. This is because the amount of pressure exerted by an electrode against the patient’s skin may significantly affect the current measurements. For example, if a patient presses against the ground electrode much harder when the probe electrode is in a first position than when the probe electrode is in a second position, the extra pressure in the first position may, unless corrective measures are taken, cause current measurements at the two positions to be incomparable.

In some implementations, this problem (different amounts of pressure exerted by the user affects magnitude of current measurements) is mitigated by employing a pressure sensor that measures the amount of force or pressure exerted against a ground electrode or probe electrode. For instance, the ground electrode or probe electrode may include or be attached to a pressure sensor. For instance, each of the six conductive pads 452, 453, 454, 480, 481, 470 of the ground electrode in FIGS. 4 and 5 may include or be attached to a pressure sensor. Any type of pressure sensor may be employed. For example, the pressure sensor may comprise: (a) a piezoresistive strain gauge; (b) a capacitive strain gauge (e.g., a variable capacitor in which capacitance decreases as a diaphragm deforms due to increasing pressure); (c) an electromagnetic pressure sensor (e.g., that measures displacement of a diaphragm by changes in inductance, or by Hall Effect, or by eddy current); (d) an optical strain gauge (e.g., that employs fiber Bragg gratings); or (e) a potentiometric strain gauge (e.g., in which change of position of a conductive element causes a change in resistance). Each time that a cross-body current is measured, the pressure sensor may measure pressure (or force) exerted against the probe electrode or the ground electrode.

In some cases, the ground electrode and probe electrode are rigid parts that are part of a rigid single structure and that thus are in a fixed position relative to each other, except for any movement that occurs due to varying displacement within a pressure sensor due to varying pressure or force exerted against the pressure sensor.

Alternatively: (a) smartphone 450 may be replaced by any other mobile computing device; and (b) rigid structure 400 may be a case that surrounds, and holds in place, the mobile computing device. For instance, the mobile computing device may be a tablet computer, notebook computer, mobile internet device, personal digital assistant, handheld PC, or ultra-mobile PC.

FIG. 6 shows a closeup view of the spring-mounted pad 491, which is part of a ground electrode. This spring-mounted pad includes a conductive tip 492, a spring 495, a rod 493, and a pressure sensor 494. For instance, pressure sensor 494 may comprise a piezoelectric, inductive, potentiometric or optical pressure sensor. Rod 493 is physically attached to conductive tip 492. Pressure exerted against conductive tip 492 causes the tip 492 and rod 493 to be displaced. Specifically, tip 492 and rod 493 are constrained to move along a single axis in a limited range of motion. Varying displacement of rod 493 is measured, as a proxy for the pressure (or force) exerted against tip 492.

Each of six conductive pads 452, 453, 454, 480, 481, 470 of the ground electrode in FIGS. 4 and 5 may be spring-mounted, in the manner shown in FIG. 6 .

In FIGS. 1, 2, 3, 4 and 5 , one or both of the ground electrode and tip of the probe electrode may comprise a metallic alloy (e.g., copper/silver) that has antibacterial and antiviral properties. Alternatively, one or both of the ground electrode and tip of the probe electrode may comprise conductive rubber.

In some implementations, the ground electrode is temporarily attached to a patient’s skin, rather than being held by a patient. For instance, the ground electrode may have an adhesive, conductive surface that adheres to the patient’s skin. In some cases: (a) the ground electrode has multiple pads, and (b) each of the pads has a sticky, conductive surface that clings to the patient’s skin.

In some alternative implementations, skin conductivity (or resistance) is measured instead of measuring current that flows through a patient’s body between two electrodes. For instance, in some implementations, skin conductivity (or resistance) is measured by an infrared or optical sensor. In some cases, the sensor that measures skin conductivity (or resistance) does not contact the patient’s skin. For example, a contact- less infrared or optical sensor may be employed to measure skin conductivity.

Measurement Locations

Before discussing measurement locations, let us first define “forearm” and “leg”. As used herein, “forearm” means the portion of an upper limb of a human that is distal to the elbow. Thus, a forearm includes: (a) a hand; (b) a wrist; and (c) a region between elbow and wrist. As used herein, “leg” means the portion of a lower limb of a human that is distal to the knee. Thus, a leg includes a crus, an ankle and a foot.

As noted above, the current sensor may measure the cross-body electrical currents while the probe electrode is positioned at 24 locations on the skin of the patient’s extremities, one location at a time. The measurement locations may consist of: (a) six locations on the right foot and six corresponding locations on the left foot; and (b) six locations on the right hand (or right wrist) and six corresponding locations on the left hand (or left wrist).

FIGS. 7, 8, and 9 show six locations 701, 702, 703, 704, 705, 706 on the right foot, at which the probe electrode may be placed (one location at a time) while the current sensor measures cross-body currents. These six locations on the right foot are positioned on acupuncture meridians. Specifically, locations 701, 702, 703, 704, 705, 706 are positioned on the Spleen, Liver, Kidney, Bladder, Gall Bladder, and Stomach acupuncture meridians, respectively. In acupuncture terminology: (a) location 701 is sometimes called SP3 or Spleen 3; (b) location 702 is sometimes called LR3 or Liver 3; (c) location 703 is sometimes called KI4 or Kidney 4; (d) location 704 is sometimes called BL65 or Bladder 65; (e) location 705 is sometimes called GB40 or Gall Bladder 40; and (f) location 706 is sometimes called ST42 or Stomach 42.

Likewise, the probe electrode may be placed (one location at a time) at six locations on the left foot, while the current sensor measures cross-body currents. These first, second, third, fourth, fifth and sixth locations on the left foot may be bilaterally symmetric with locations 701, 702, 703, 704, 705, and 706, respectively, on the right foot. Put differently, these first, second, third, fourth, fifth and sixth locations on the left foot of a patient may have reflectional symmetry (about the patient’s sagittal plane) with locations 701, 702, 703, 704, 705, and 706, respectively, on the right foot of the patient. These six locations on the left foot may be positioned on the same acupuncture meridians—- and have the same acupuncture meridian point numbers - as the respective corresponding locations on the right foot. For instance, the location on the left foot that is bilaterally symmetric with location 701 may be on the Spleen acupuncture meridian and may also be called SP3 or Spleen 3.

FIGS. 10A and 10B show six locations 801, 802, 803, 804, 805, 806 on the right forearm, at which the probe electrode may be placed (one location at a time) while the current sensor measures cross-body currents. These six locations on the right forearm are positioned on acupuncture meridians. Specifically, locations 801, 802, 803, 804, 805, 806 are positioned on the Lung, Pericardium, Heart, Small Intestine, Triple Heater and Large Intestine acupuncture meridians, respectively. In acupuncture terminology: (a) location 801 is sometimes called LU9 or Lung 9; (b) location 802 is sometimes called PC7 or Pericardium 7; (c) location 803 is sometimes called HT7 or Heart 7; (d) location 804 is sometimes called SI5 or Small Intestine 5; (e) location 805 is sometimes called TH4 or Triple Heater 4; and (f) location 806 is sometimes called LI5 or Large Intestine 5.

Likewise, the probe electrode may be placed (one location at a time) at six locations on the left forearm, while the current sensor measures cross-body currents.

These first, second, third, fourth, fifth and sixth locations on the left forearm may be bilaterally symmetric with locations 801, 802, 803, 804, 805, and 806, respectively, on the right forearm. Put differently, these first, second, third, fourth, fifth and sixth locations on the left forearm of a patient may have reflectional symmetry (about the patient’s sagittal plane) with locations 801, 802, 803, 804, 805, and 806, respectively, on the right forearm of the patient. These six locations on the left forearm may be positioned on the same acupuncture meridians—- and have the same acupuncture point numbers - as the respective corresponding locations on the right forearm. For instance, the location on the left forearm that is bilaterally symmetric with location 801 may be on the Lung acupuncture meridian and may also be called LU or Lung 3.

As used herein, “Prototype Measurement Locations” means the 24 locations that are mentioned in the preceding four paragraphs (i.e., twelve locations 701, 702, 703, 704, 705, 706, 801, 802, 803, 804, 805, and 806 on the right side of a patient and twelve bilaterally symmetric locations on the left side of a patient).

Alternatively, the probe electrode may be placed at other acupuncture points. For each Prototype Measurement Location, another acupuncture point on the same meridian may be used instead. Put differently, rather than place the probe electrode at a Prototype Measurement Point on a given meridian, the probe electrode may instead be placed on another acupuncture point on the same meridian. For example, rather than place the probe electrode at a Prototype Measurement Point that is on a forearm and on a given meridian, the probe electrode may instead be placed on another acupuncture point that is on the same forearm and on the same meridian. Likewise, rather than place the probe electrode at a Prototype Measurement Point that is on a given meridian and is distal to a knee, the probe electrode may instead be placed on another acupuncture point that is on the same meridian and is distal to the same knee.

For instance: (a) to measure a cross-body electrical current for a Spleen meridian, the probe electrode may be placed at an acupuncture point that is on the Spleen meridian and is distal to a knee (e.g., at any of Spleen meridian points SP1 to SP8, inclusive); (b) to measure a cross-body electrical current for a Liver meridian, the probe electrode may be placed at an acupuncture point that is on the Liver meridian and is distal to a knee (e.g., at any of Liver meridian points LR1 to LR6, inclusive); (c) to measure a cross-body electrical current for a Kidney meridian, the probe electrode may be placed at an acupuncture point that is on the Kidney meridian and is distal to a knee (e.g., at any of Kidney meridian points KI1 to KI9, inclusive); (d) to measure a cross-body electrical current for a Bladder meridian, the probe electrode may be placed at an acupuncture point that is on the Bladder meridian and is distal to a knee (e.g., at any of Bladder meridian points BL55 to BL67, inclusive); (e) to measure a cross-body electrical current for a Gall Bladder meridian, the probe electrode may be placed at an acupuncture point that is on the Gall Bladder meridian and is distal to a knee (e.g., at any of Gall Bladder meridian points GB35 to GB44, inclusive); (f) to measure a cross-body electrical current for a Stomach meridian, the probe electrode may be placed at an acupuncture point that is on the Stomach meridian and is distal to a knee (e.g., at any of Stomach meridian points ST36 to ST45, inclusive); (g) to measure a cross-body electrical current for a Lung meridian, the probe electrode may be placed at an acupuncture point that is on the Lung meridian and is distal to an elbow (e.g., at any of Lung meridian points LU6 to LU11, inclusive); (h) to measure a cross-body electrical current for a Pericardium meridian, the probe electrode may be placed at an acupuncture point that is on the Pericardium meridian and is distal to an elbow (e.g., at any of Pericardium meridian points PC4 to PC9, inclusive); (i) to measure a cross-body electrical current for a Heart meridian, the probe electrode may be placed at an acupuncture point that is on the Heart meridian and is distal to an elbow (e.g., at any of Heart meridian points HT4 to HT9, inclusive); (j) to measure a cross-body electrical current for a Small Intestine meridian, the probe electrode may be placed at an acupuncture point that is on the Small Intestine meridian and is distal to an elbow (e.g., at any of Small Intestine meridian points SI 1 to SI 7, inclusive); (k) to measure a cross-body electrical current for a Triple Heater meridian, the probe electrode may be placed at an acupuncture point that is on the Triple Heater meridian and is distal to an elbow (e.g., at any of Triple Heater meridian points TH1 to TH9, inclusive); and (1) to measure a cross-body electrical current for a Large Intestine meridian, the probe electrode may be placed at an acupuncture point that is on the Large Intestine meridian and is distal to an elbow (e.g., at any of Large Intestine meridian points LI 1 to LI 9, inclusive).

Alternatively, in some implementations, less than 24 measurement locations are employed in a single diagnostic session. For instance, in some cases, the probe electrode is positioned (at different times during a single diagnostic session) at a total of 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11 or 12 locations, while the current sensor measures cross-body electrical currents. In some cases: (a) the probe electrode is placed at 12 or less measurement locations during a single diagnostic session; and (b) half of the locations are on a right limb and half are on a left limb in bilaterally symmetric locations. In some cases, during a single diagnostic session, the probe electrode is placed at 12 or less measurement locations that are all on one or two forearms of the patient. For instance, in some cases, during a single diagnostic session, the probe electrode is placed at only 12 or less locations, all of which are on one or two wrists of the patient.

In some alternative implementations of this invention, the measurement locations are not on acupuncture points and are not located on acupuncture meridians. Put differently, when taking measurements of cross-body currents, the probe electrode may be pressed against the patient’s skin at locations that are not acupuncture points and that are not on acupuncture meridians.

Currents

As used herein, a “Prototype Current” means an electrical current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of another limb of the patient at a Prototype Measurement Location.

As used herein, an “SP current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a patient; and (b) the probe electrode is touching skin of a leg of the patient at a location on the Spleen acupuncture meridian. As a non- limiting example, the location mentioned in the preceding sentence may be a Spleen 3 acupuncture point (e.g., location 701 on the patient’s right leg in FIG. 7 or a bilaterally symmetric location on the patient’s left leg).

As used herein, an “LR current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a patient; and (b) the probe electrode is touching skin of a leg of the patient at a location on the Liver acupuncture meridian. As a non- limiting example, the location mentioned in the preceding sentence may be a Liver 3 acupuncture point (e.g., location 702 on the patient’s right leg in FIG. 8 or a bilaterally symmetric location on the patient’s left leg).

As used herein, a “KI current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a patient; and (b) the probe electrode is touching skin of a leg of the patient at a location on the Kidney acupuncture meridian. As a non- limiting example, the location mentioned in the preceding sentence may be a Kidney 4 acupuncture point (e.g., location 703 on the patient’s right leg in FIG. 7 or a bilaterally symmetric location on the patient’s left leg).

As used herein, a “BL current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a patient; and (b) the probe electrode is touching skin of a leg of the patient at a location on the Bladder acupuncture meridian. As a non- limiting example, the location mentioned in the preceding sentence may be a Bladder 65 acupuncture point (e.g., location 704 on the patient’s right leg in FIG. 9 or a bilaterally symmetric location on the patient’s left leg).

As used herein, a “GB current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a patient; and (b) the probe electrode is touching skin of a leg of the patient at a location on the Gall Bladder acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Gall Bladder 40 acupuncture point (e.g., location 705 on the patient’s right leg in FIG. 9 or a bilaterally symmetric location on the patient’s left leg).

As used herein, an “ST current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a patient; and (b) the probe electrode is touching skin of a leg of the patient at a location on the Stomach acupuncture meridian. As a non- limiting example, the location mentioned in the preceding sentence may be a Stomach 42 acupuncture point (e.g., location 706 on the patient’s right leg in FIG. 8 or a bilaterally symmetric location on the patient’s left leg).

As used herein, an “LU current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of the opposite forearm of the patient at a location on the Lung acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Lung 9 acupuncture point (e.g., location 801 on the patient’s right forearm in FIG. 10A or a bilaterally symmetric location on the patient’s left forearm).

As used herein, a “PC current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of the opposite forearm of the patient at a location on the Pericardium acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Pericardium 7 acupuncture point (e.g., location 802 on the patient’s right forearm in FIG. 10A or a bilaterally symmetric location on the patient’s left forearm).

As used herein, an “HT current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of the opposite forearm of the patient at a location on the Heart acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Heart 7 acupuncture point (e.g., location 803 on the patient’s right forearm in FIG. 10A or a bilaterally symmetric location on the patient’s left forearm).

As used herein, an “SI current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of the opposite forearm of the patient at a location on the Small Intestine acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Small Intestine 5 acupuncture point (e.g., location 804 on the patient’s right forearm in FIG. 10B or a bilaterally symmetric location on the patient’s left forearm).

As used herein, a “TH current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of the opposite forearm of the patient at a location on the Triple Heater acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Triple Heater 4 acupuncture point (e.g., location 805 on the patient’s right forearm in FIG. 10B or a bilaterally symmetric location on the patient’s left forearm).

As used herein, an “LI current” means an electric current between a probe electrode and a ground electrode, which current is measured while: (a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of the opposite forearm of the patient at a location on the Large Intestine acupuncture meridian. As a non-limiting example, the location mentioned in the preceding sentence may be a Large Intestine 5 acupuncture point (e.g., location 806 on the patient’s right forearm in FIG. 10B or a bilaterally symmetric location on the patient’s left forearm).

As used herein: (a) “right side” of a patient means the portion of the patient’s body to the right of the patient’s sagittal plane; (b) “left side” of a patient means the portion of the patient’s body to the left of the patient’s sagittal plane; (c) a current “on the right side” means a current that is measured while the probe electrode is placed at a measurement location on a patient’s right side; and (d) a current “on the left side” means a current that is measured while the probe electrode is placed at a measurement location on a patient’s right side. To say that a current is in a specific current range “on both sides” means that the current is in the specific current range when measured while the probe electrode is positioned at a measurement location on the right side and also in the same current range when measured while the probe electrode is positioned at a bilaterally symmetric location on the left side.

FIG. 11 shows cross-body currents, in an illustrative implementation of this invention. In FIG. 11 , sagittal plane 1121 divides a patient’s body into right and left sides. Transverse plane 1120 intersects the patient’s navel and divides the patient’s body into upper and lower halves.

In FIG. 11 , a first cross-body electric current flows between: (a) a probe electrode that touches the patient’s left foot at location 1102; and (b) a ground electrode that touches the patient’s right palm at location 1101. This first current passes through both sagittal plane 1121 and transverse plane 1120.

In FIG. 11 , a second cross-body electric current flows between: (a) a probe electrode that touches the patient’s right foot at location 1103; and (b) a ground electrode that touches the patient’s right palm at location 1101. This second current passes through transverse plane 1120.

In FIG. 11 , a third cross-body electric current flows between: (a) a probe electrode that touches the patient’s left forearm at location 1104; and (b) a ground electrode that touches the patient’s right palm at location 1101. This third current passes through sagittal plane 1121 and, depending on the positions of the user’s hands, may also pass through transverse plane 1120.

Current Ranges

In some implementations, during a single diagnostic session, the current sensor (e.g., 122) takes multiple measurements of electrical current at each measurement location. Put differently, the current sensor may take multiple measurements of electrical current at each point on the patient’s skin where the probe electrode is placed.

Each of these current measurements may be calibrated. For instance, the current measurements may be calibrated based on simultaneous pressure measurement(s) that is/are indicative of pressure or force exerted against the probe electrode or ground electrode. The calibration may eliminate the impact of varying pressure or force on the magnitude of the current readings.

The calibrated measurements for a single measurement location may be filtered to eliminate outliers.

Thus, for each single measurement location, multiple calibrated, filtered current measurements may be taken.

The measurements for a single measurement location may (after any calibration and/or filtering) be averaged, to yield an average value for that measurement location. For instance, the current sensor may take 20 measurements of SP current while:

-   (a) the ground electrode is touching the same region of skin on a     hand of a patient; and -   (b) the probe electrode is touching the patient’s skin at location     701 on the patient’s right leg. These 20 measurements of SP current     may (after any calibration and/or filtering) be averaged, resulting     in an average SP current.

This process of calculating an average current for each measurement location may be repeated for multiple measurement locations on a single patient during a single diagnostic session. In some cases: (a) current measurements are taken at 24 different measurement locations on a single patient during a single diagnostic session; and (g) 24 average currents are calculated, one for each of the 24 measurement locations.

The average values for the respective measurement locations for the diagnostic session may then be averaged, resulting in an overall average current for the patient for the diagnostic session. For instance, in some cases: (a) there are 24 measurement locations; and (b) the overall average current is an average of the 24 average currents for the respective 24 measurement locations.

A set of current ranges for the patient for the diagnostic session may then be calculated. In some cases, we call these current ranges: (a) “way above average”; (b) “above average”; (c) “average”; (d) “below average” and (e) “way below average”. The amperage in the “way above average” range is greater than in the “above average” range, which in turn is greater than in the “average” range, which in turn is greater than in the “below average” range, which in turn is greater than in the “way below average” range.

In some implementations: (a) the “average range” is selected in such a way as to be centered on the overall average current; and (b) the magnitude of the difference (in amperes) between the lower bound of the “way above average” range and the upper bound of the “average” range is equal to the magnitude of the difference (in amperes) between the lower bound of the “average” range and the upper bound of the “way below average” range.

In some implementations, one or more computers calculate, based on the overall average current for a patient for a diagnostic session, what we call Prototype Current Ranges for the patient for the diagnostic session.

As used herein, “Prototype Current Ranges” for a diagnostic session mean a set of current ranges in which: (a) the set consists of five current ranges, specifically, a “way above average” range, an “above average” range, an “average” range, a “below average” range, and a “way below average” range; (b) amperage in the way above average range is greater than amperage in the above average range, which in turn is greater than amperage in the average range, which in turn is greater than amperage in the below average range, which in turn is greater than amperage in the way below average range; (c) the upper bound of the average range is equal to the overall average current for the diagnostic session plus 25 microamps; (d) the lower bound of the average range is equal to the overall average current for the diagnostic session minus 25 microamps; (e) the upper bound of the above average range is equal to the overall average current for the diagnostic session plus 50 microamps; and (e) the lower bound of the below average range is equal to the overall average current minus 50 microamps. Notwithstanding the foregoing, a current range in the “Prototype Current Ranges” shall be truncated or eliminated to the extent needed to cause all values in the Prototype Current Ranges to be positive. For purposes of the definition of “Prototype Current Ranges” for a diagnostic session, the “overall average current” means an average of currents (after any calibration and/or filtering) for the respective measurement locations during the diagnostic session.

In illustrative implementations, a current for each measurement location is assigned to one of the calculated current ranges. For instance, in some cases:

(a) current measurements are taken at 24 different measurement locations on a single patient during a single diagnostic session; (b) 24 currents are calculated, one for each of the 24 measurement locations; and (c) each of the 24 currents is assigned to one of the calculated current ranges. Each current that is assigned to a current range may itself be an average, calibrated and/or filtered current, as described above.

Alternatively, in some cases: (a) only a single current measurement is taken at each measurement location during a diagnostic session; (b) the overall average current is equal to the average of these single current measurement for the respective measurement locations; and (c) the single current measurements for the respective measurement locations are each assigned to a current range. In some cases, calibration and/or filtering are not performed, and the overall average current is calculated with uncalibrated and/or unfiltered data.

Lookup Table

In some implementations, after currents for the respective measurement locations are each assigned to a current range, a computer employs a look up table to determine one or more medical conditions that are indicated by one or more of these currents. For instance, the computer may determine whether one or more of these currents is or are in a specific state that is listed in the lookup table, and may further determine that this specific state is associated (by the look up table) with one or more specific medical conditions, and may thus conclude that the current readings in the diagnostic session indicate that these one or more specific medical conditions are present. Put differently, the computer may conclude that the specific state (of electrical currents) exists and is a biomarker for the one or more specific medical conditions.

For instance, a computer: (a) may determine that HT current and PC current are in a specific state, in which HT current is below average on the left, the right or both sides of a patient and the PC current is below average on the left, the right or both sides of the patient; (b) may access a lookup table and determine that this specific state is associated (by the lookup table) with coronary artery disease; and (b) may thus conclude that coronary artery disease is indicated by the current readings. Put differently, the computer may conclude that a specific current state (of HT and PC currents) exists and is a biomarker for coronary artery disease.

In some use scenarios, a single current state is associated (by the lookup table) with more than one medical conditions.

Each specific current state may consist of either: (a) a current range for a single current (e.g., SP current is below average on left and right sides); or (b) current ranges for multiple respective currents (e.g., GB current is way below average on left and right sides and LR current is average on left and right sides).

In some implementations, the lookup table is also employed to determine a confidence level or probability for a particular medical condition. For instance, the lookup table may associate a specific current state with both: (a) a medical condition; and (b) a confidence level or probability for that condition. The confidence level or probability may be explicit or implicit. For instance, the lookup table may indicate that a recommendation should be made for further medical testing, in order to evaluate whether or not a specific medical condition is actually present. We sometimes call this a “rule- out” recommendation.

In some implementations, the lookup table includes all or part of the information set forth in Table 1 below. For instance, the lookup table may include at least part of the information (about medical conditions, electric current states and associations between electrical current state and medical condition) which is set forth in Table 1 below.

TABLE 1 Class Medical Condition Electrical Current State 1. B anemia (a)SP current is below average on left and right sides; and (b)KI current is average on left and right sides; and (c)BL current is average on left and right sides. 2. A anxiety [(a) HT current is above average on left, right or both sides; and (b) LR current is average on left and right sides] and/or [(a) LU current is average on left, right or both sides; and (b) PC current is average on left, right or both sides; and (c) HT current is average on left, right or both sides; and (d) LR current is average on left and right sides.] and/or [(a) HT current is above average on left and right sides; and (b) SI current is above average on left, right or both sides]. 3. D back injury BL current is above average on left, right or both sides. 4. D back injury -with nerve damage and no pain (a)BL current is below average on left, right or both sides; and (b)SP current is below average on left, right or both sides. 5. D back injury -with nerve damage and pain (a)BL current is above average on left, right or both sides; and (b)SP current is below average on left, right or both sides. 6. D back injury -with pain BL current is above average on left, right or both sides. 7. B bacterial infection (a)SP current is way above average on left and right sides; and (b)KI current is average on left and right sides; and (c)BL current is average on left and right sides. 8. A bipolar disorder - in depressive state (a)PC current is below average on left and right sides; and (b)HT current is below average on left and right sides; and (c)LR current is above average on left and right sides. 9. A bipolar disorder - in manic state (a) LU current is way above average on left and right sides; and (b) PC current is way above average on left and right sides; and (c) HT current is way above average on left and right sides; and (d) LR current is way above average on left and right sides. 10. G bladder dysfunction (a)BL current is above average on left, right or both sides; and/or (b)BL current is below average on left, right or both sides. 11. B bladder infection (a)KI current is way above average on left and right sides; and (b)BL current is way above average on left and right sides; and (c)SP current is way above average on left and right sides. 12. I blood pressure medication overdose (a)PC current is below average on left and right sides; and (b)HT current is below average on left and right sides. 13. G chronic fatigue and/or chronic sleep deficit (a)HT current is below average on left, right or both sides; and (b)PC current is below average on left, right or both sides. 14. B chronic obstructive pulmonary disease (a)LU current is above average on left and right sides; xor (b)LU current is below average on left and right sides. 15. B constipation (a)LI current is below average on left, right or both sides; and (b)SI current is average on left, right or both sides; and (c)TH current is average on left, right or both sides. 16. M coronary artery disease (a)HT current is below average on left, right or both sides; and (b)PC current is below average on left, right or both sides. 17. J deficiency in caloric food intake (a)SI current is way below average on left and right sides; and (b)TH current is way below average on left and right sides; and (c)LI current is way below average on left and right sides; and (d)ST current is way below average on left and right sides; and (e)GB current is way below average on left and right sides. 18. J deficiency in protein intake LR current is below average on left and right sides. 19. I mild (i.e., non- clinical) depression (a)HT current is above average or below average on left side; and (b)HT current is above average or below average on right side; and (c)PC current is above average or below average on left side; and (d)PC current is above average or below average on right side; and (e)LR current is above average on left and right sides. 20. A clinical depression (a)PC current is below average on left, right or both sides; and (b)HT current is below average on left, right or both sides; and (c)LR current is above average on left, right or both sides; and (d)ST current is average or below average on left, right or both sides. 21. K poor glycemic control (if patient has diabetes mellitus) SP current is below average on left and right sides. 22. I digestive disorder (a) LI current is average or below average on left side; and (b) LI current is average or below average on right side; and (c)ST current is way below average on left and right sides; and (d)GB current is average on left and right sides. 23. F duodenal irritation (a) SI current is above average on left, right or both sides; and (b) [(HT current is average on left and right sides) xor (HT current is below average on left and right sides)]; and (c) [(LR current is average on left and right sides) xor (LR current is below average on left and right sides)]; and (d) LI current is average on left and right sides. 24. L dysautonomia (a)PC current is way below average on left and right sides; and (b)HT current is way below average on left and right sides; and (c)SP current is way below average on left and right sides; and (d)KI current is way below average on left and right sides; and (e)BL current is way below average on left and right sides. 25. L dysphagia (a) [(PC current is above average on left and right sides) xor (PC current is way above average on left and right sides)]; and (b) GB current is above average on left and right sides. 26. J excessive fat intake (a)LR current is way above average on left and right sides; and (b)GB current is above average or way above average on left side; and (c)GB current is above average or way above average on right side; and (d)ST current is above average or way above average on left side; and (e)ST current is above average or way above average on right side. 27. I fatigue (a)PC current is below average on left, right or both sides; and (b)HT current is below average on left, right or both sides. 28. I food-related sinus allergy (a)LU current is below average on left, right or both sides; and (b)LR current is above average on left, right or both sides. 29. G gallbladder disorder (a)GB current is way below average on left and right sides; and (b)LR current is average on left and right sides. 30. M gastroparesis (a)ST current is way below average on left and right sides; and (b)GB current is average on left and right sides; and (c)SP current is way below average on left, right or both sides; and (d)KI current is way below average on left, right or both sides; and (e)BL current is way below average on left, right or both sides. 31. G headaches (a)PC current is below average on left, right or both sides; and (b)HT current is below average on left, right or both sides; and (c)[(KI current is above average on left, right or both sides) and/or (SP current is above average on left, right or both sides)]. 32. C hyper-metabolism (a)SI current is way above average on left and right sides; and (b)TH current is way above average on left and right sides; and (c)LI current is way above average on left and right sides. 33 . C hypertension (a)PC current is above average on left and right sides; and (b)[(HT current is above average on left and right sides) xor (HT current is way above average on left and right sides)]. 34. C hyperthyroid (a)TH current is above average on left and right sides; and (b)SI current is average on left and right sides; and (c)LI current is average on left and right sides. 35. C hypotension (a)[(PC current is below average on left and right sides) xor (PC current is way below average on left and right sides)]; and (b)[(HT current is below average on left and right sides) xor (HT current is way below average on left and right sides)]. 36. C hypothyroid (a)TH current is below average on left and right sides; and (b)SI current is average on left and right sides; and (c)LI current is average on left and right sides. 37. K incontinence (if patent is female) (BL current is below average on left, right or both sides) and/or (BL current is above average on left, right or both sides)]. 38. Q inflammation (a)LR current is way above average on left and right sides; and (b)GB current is average on left and right sides; and (c)ST current is average on left and right sides. 39. F irritation of stomach lining ST current is way above average on left, right or both sides. 40. E irritation of nerves in both upper back and lower back (a)BL current is above average on left, right or both sides; and (b)KI current is above average on left, right or both sides. 41. E irritation of nerves in lower back BL current is above average on left and right sides. 42. E irritation of nerves in neck KI current is above average on left, right or both sides. 43. M kidney failure (a)KI current is below average on left and right sides; and (b)BL current is below average on left and right sides. 44. I large intestine disorder (with symptoms other than or in addition to large intestine irritation) (a)LI current is below average on left, right or both sides; and/or (b)LI current is above average on left, right or both sides. 45. F large intestine irritation (a)LI current is way above average on left, right or both sides; and (b)SI current is average on left and right sides. 46. D lower back injury BL current is above average on left, right or both sides. 47. D lower back pain BL current is above average on left, right or both sides. 48. Q lung cancer LU current is way above average on left, right or both sides. 49. Q lung disease (excluding lung cancer) (a)LU current is above average on left, right or both sides; and/or (b)LU current is below average on left, right or both sides. 50. K menstruating (if patient is a woman) (a)SP current is above average on left and right sides; and (b)BL current is above average on left and right sides; and (c)KI current is above average on left and right sides. 51. Q Microcirculatory disease PC current is below average on left, right or both sides. 52. N Microcirculatory -orthostatic hypotension (a)PC current is below average on left and right sides; and (b)HT current is below average on left and right sides; and (c)SP current is below average on left and right sides; and (d)KI current is below average on left and right sides; and (e)BL current is below average on left and right sides. 53. N multiple sclerosis (a)KI current is below average on left and right sides; and (b)SP current is below average on left and right sides; and (c)BL current is below average on left and right sides. 54. H nerve damage in lower back BL current is below average on left, right or both sides. 55. H nerve damage in neck KI current is below average on left, right or both sides. 56. H nerve damage due to fall on tailbone (in female patient) (a)BL current is below average on left and right sides; and (b)KI current is average on left and right sides; and (c)SP current is average on left and right sides. 57. H nerve damage in upper back KI current is below average on left, right or both sides. 58. N benign prostatic hyperplasia (a)BL current is below average on left, right or both sides; and (b)KI current is average on left, right or both sides; and (c)SP current is average on left, right or both sides. 59. Q prostate cancer (a)BL current is above average on left, right or both sides; and (b)KI current is average on left and right sides; and (c)SP current is average on left and right sides. 60. P reflux (a)GB current is above average on left, right or both sides; and (b)ST current is above average on left, right or both sides. 61. P sinus infection (a)SP current is way above average on left and right sides; (b)KI current is average on left and right sides; and (c)BL current is average on left and right sides; and (c) LU current is way below average on left and right sides. 62. L sleep disorder (a)HT current is below average on left, right or both sides; and (b)PC current is below average on left, right or both sides. 63. I impaired stomach motility (a)ST current is way below average on left and right sides; and (b)GB current is average on left and right sides. 64. J excessive stress [(a) HT current is above average on left, right or both sides; and (b) PC current is above average on left, right or both sides; and (c) SI current is below average on left, right or both sides] and/or [(a) HT current is average on left, right or both sides; and (b) PC current is average on left, right or both sides; and (c) ((SI current is above average on left, right or both sides) and/or (SI current is way above average on left, right or both sides))]. 65. P viral infection (a)SP current is way below average on left and right sides; and (b)KI current is average or way above average on left, right or both sides; and (c)BL current is average or way above average on left, right or both sides.

Table 1 has 65 rows.

Table 1 lists 65 electrical current states, i.e., one electrical current state per row. As used herein, “Prototype Electrical Current State” means an electrical current state that is listed in a row of Table 1. For instance, the Prototype Electrical Current State listed in row 3 of Table 1 is “BL current is below average on left, right or both sides.” Also, for instance, the Prototype Electrical Current State listed in row 4 of Table 1 is “(a) BL current is below average on left, right or both sides; and (b) SP current is below average on left, right or both sides.”

Each current range listed in Table 1 is a Prototype Current Range. Specifically, each time that a current range “way above average”, “above average”, “average”, “below average” or “way below average” is listed in Table 1, that current range is a Prototype Current Range. For instance, in row 3 of Table 1, “above average” is a Prototype Current Range. Also, for instance, in row 28 of Table 1, “below average” and “above average” are each a Prototype Current Range.

Table 1 lists 65 medical conditions; i.e., one medical condition per row. As used herein, “Prototype Medical Condition” means a medical condition listed in a row of Table 1. For instance, the Prototype Medical Conditions listed in rows 1, 2 and 65 of Table 1 are anemia, anxiety and viral infection, respectively.

In each row in Table 1, the electrical current state listed in that row indicates that the patient has the medical condition listed in that row. Put differently, in each row in Table 1, the electrical current state listed in that row is a biomarker for the medical condition listed in that row. Likewise, in each row in Table 1, the electrical current state listed in that row is a factor that, in a differential diagnosis, points toward (or weighs in favor of) concluding that the patient has at least the medical condition listed in that row.

Table 1 associates Prototype Medical Conditions with respective Prototype Electrical Current States. Specifically, Table 1 associates the Prototype Medical Condition listed in each row of Table 1 with the Prototype Electrical Current State listed in that row. As a non-limiting example, Table 1 associates the Prototype Medical Condition listed in row 48 of Table 1 (i.e., lung cancer) with the Prototype Electrical Current State listed in row 48 of Table 1 (i.e., “LU current is way above average on left, right or both sides”.)

As used herein, when the first letter of the verb “Associate” is capitalized, then to “Associate” means to associate, by a lookup table, a Prototype Medical Condition listed in a row of Table 1 with the Prototype Electrical Current State listed in that row of Table 1. The term “Associate” does not require accessing Table 1 itself; instead “Associate” requires a lookup table to make the same association as is made in a row of Table 1. For instance, if a lookup table were to associate lung cancer with the Prototype Electrical Current State “LU current is way above average on left, right or both sides”, then the lookup table would be Associating lung cancer with that Prototype Electrical Current State. (This is because row 48 of Table 1 makes that association) . The definition of “Associate” in this paragraph does not create any implication regarding the meaning of the word “associate” when the first letter of the word is not capitalized.

In each row in Table 1, if the electrical current state for that row does not explicitly mention a specific current, then that specific current may be in any Prototype Current Range. For instance: (a) in row 1 of Table 1, only SP current, KI current and BL current are explicitly mentioned; and (b) in the electrical current state listed in row 1, other currents (e.g., LR, GB, ST, LU, PC, HT, SI, TH and LI currents) may be in any Prototype Current Range.

Each current that is on a particular side of a patient and that is listed in Table 1 may have a value derived from: (a) a single measurement at a particular measurement location (after any calibration) or (b) multiple measurements at the particular measurement location (after any calibration and filtering). If a current listed in Table 1 has a value that is derived from multiple measurements at a particular measurement location, then that value is an average of the multiple measurements (after any calibration and filtering).

In some use scenarios, each current listed in Table 1 is a Prototype Current that is measured when a probe electrode is touching a Prototype Measurement Location. Likewise, in some use scenarios: (a) each current that is on a specific side of a patient and that is listed in Table 1 is an electric current between a probe electrode and a ground electrode, which current is measured while:

(a) the ground electrode is touching skin of a hand of a forearm of a patient; and (b) the probe electrode is touching skin of another limb of the patient at a Prototype Measurement Location on that specific side of the patient.

This invention may be employed to accurately detect and diagnose medical conditions. For instance, in some implementations of this invention: (a) the Prototype Currents listed in Table 1 are measured while the probe electrode is placed at the Prototype Measurement Locations; (b) the measured currents are assigned to Prototype Current Ranges; and (c) accurate diagnoses of medical conditions are made based on the respective associations (between electrical current states and medical conditions) that are set forth in Table 1.

In Table 1, each medical condition is assigned a class. Specifically, each medical condition listed in a row of Table 1 is classified as being in a particular class, which particular class is listed in that row. For instance, in row 1 of Table 1, the medical condition of anemia is classified as being in Class B.

As used herein: (a) “Class A Condition” means a medical condition that is, in Table 1, classified as being in Class A; (b) “Class B Condition” means a medical condition that is, in Table 1, classified as being in Class B; (c) “Class C Condition” means a medical condition that is, in Table 1, classified as being in Class C; (d) “Class D Condition” means a medical condition that is, in Table 1, classified as being in Class D; (e) “Class E Condition” means a medical condition that is, in Table 1, classified as being in Class E; (f) “Class F Condition” means a medical condition that is, in Table 1, classified as being in Class F; (g) “Class G Condition” means a medical condition that is, in Table 1, classified as being in Class G; (h) “Class H Condition” means a medical condition that is, in Table 1, classified as being in Class H; (i) “Class I Condition” means a medical condition that is, in Table 1, classified as being in Class I; (j) “Class J Condition” means a medical condition that is, in Table 1, classified as being in Class J; (k) “Class K Condition” means a medical condition that is, in Table 1, classified as being in Class K; (1) “Class L Condition” means a medical condition that is, in Table 1, classified as being in Class L; (m) “Class M Condition” means a medical condition that is, in Table 1, classified as being in Class M; (n) “Class N Condition” means a medical condition that is, in Table 1, classified as being in Class N; (p) “Class P Condition” means a medical condition that is, in Table 1, classified as being in Class P; and (q) “Class Q Condition” means a medical condition that is, in Table 1, classified as being in Class Q. The medical conditions listed in this paragraph are each an example of a Prototype Medical Condition.

As noted above, the diagnostic system may determine whether or not a patient has a viral infection and whether or not a patient has a bacterial infection, based on electrical current measurements that take only a few minutes. This ability to quickly and accurately detect and differentiate between viral and bacterial infections enables the diagnostic system to be used as a mass-scale, rapid screening tool in a viral or bacterial epidemic.

For instance, the medical conditions listed in rows 7 and 65 of Table 1 are bacterial infection and viral infection, respectively. Table 1 associates bacterial infection with the electric current state listed in row 7 of Table 1 and associates viral infection with the electric current state listed in row 65 of Table 1. For example, if the electrical current state listed in row 65 of Table 1 is detected, then the diagnostic system may output a diagnosis that the patient has a viral infection. Likewise, if the electrical current state listed in row 7 of Table 1 is detected, then the diagnostic system may output a diagnosis that the patient has a bacterial infection.

Machine Learning

In some implementations of this invention, a computer employs a trained machine learning model instead of a lookup table, in order to predict a medical condition based on measurements of cross-body electrical currents.

In some implementations, the input to the machine learning model is data representing measurements of cross-body currents at multiple different measurement locations for a single patient during a single diagnostic session. For instance, the input to the machine learning model may comprise measurements of electrical currents, where:(a) the currents flow between a probe electrode and a ground electrode; and (b) the measurements are taken during a single diagnostic session while the patient holds the ground electrode and while the probe electrode is pressed against the patient’s skin at each of multiple different locations on limbs of the patient, one location at a time.

In some cases, the data is calibrated (e.g., to adjust for the effect, if any, of pressure exerted against an electrode) and filtered (e.g., to remove outliers) before being fed as input into the machine learning model. In some cases: (a) multiple current measurements are taken at each measurement location; (b) the multiple measurements for each given location are averaged and the resulting average current for that given location is fed as an input into the machine learning model. In some cases: (a) the currents for the respective measurement locations are assigned into current ranges; and (b) the current ranges for the respective measurement locations are fed as inputs into the machine learning model. In some cases, one or more other features are extracted from the current measurements (and/or from contextual information), and are also fed as input into the machine learning algorithm.

In some implementations, the machine learning model that is used to predict medical conditions is a supervised learning algorithm, such as a decision tree algorithm, random forests algorithm, ANN (artificial neural network), CNN (convolutional neural network), RNN (recurrent neural network), RNN with LSTM (long short term memory), RNN with Gated Recurrent Unit, MLP (multi-layered perceptron), or SVM (support vector machine) algorithm or a classifier such as a KNN (k-nearest neighbors) or naive Bayes algorithm. The supervised learning model may be trained on a training dataset that has been labeled by a health care worker or other human expert. The labels may be medical conditions. The data that is labeled may comprise electrical current measurements or data or features derived therefrom. In some cases: (a) there are practical difficulties in obtaining a sufficiently large dataset for training; and (b) a generative model (e.g., a variable autoencoder or generative adversarial network) is employed to generate a synthetic database. This synthetic database may be added to a database derived from actual measurements, in order to form a large training database for supervised learning.

In some other implementations of this invention, the machine learning model that is used to predict medical conditions is a reinforcement learning algorithm (such as a Monte Carlo, Q-learning, state-action-reward-state-action, or deep Q network algorithm). Alternatively, the machine learning model that is used to predict medical conditions is an unsupervised machine learning algorithm, such as an AE (auto-encoder), SAE (stacked auto-encoder) VAE (variational auto-encoder), DBN (deep belief network), GAN (generative adversarial network), conditional GAN, or infoGAN algorithm. Or, for instance, the machine learning model may comprise a restricted Boltzmann machine.

In some implementations, the machine learning model outputs both: (a) one or more predicted medical conditions; and (b) a confidence level or probability for each of the one or more predicted medical conditions. Again, the confidence level or probability may be explicitly stated or may be implicit. For instance, the machine learning model may output a list of medical conditions, ranked from most probable to less probable. Or, for instance, the machine learning algorithm may output a “rule-out” recommendation - that is, a recommendation that further medical tests be performed to evaluate whether or not a specific medical condition is actually present.

In some implementations: (a) the machine learning model is a supervised learning algorithm; (b) after the model is initially trained, additional data is gathered based on ongoing experiences with patients; and (c) this additional data is labeled and used for additional training of the machine learning model.

User Interface

In some implementations, one or more computers control input/output (I/O) devices in such a way as to present a GUI (graphical user interface) or audiovisual UI (user interface) to a patient, health care worker or other user. For instance, a touch screen or other electronic display screen (e.g., 133, 451) may render a GUI. The patient, health care worker or other user may interact with the GUI by inputting instructions or data via one or more I/O devices such as a touch screen, keyboard 134, or mouse 135. In some implementations, the I/O devices present an audiovisual UI, including audio information outputted by speaker 132. In this audiovisual UI, audio input by the user may be detected by microphone 131 or by a microphone onboard smartphone 450.

The GUI or UI may present (to a patient, health care worker or other user) information about, among other things: (a) electrical current measurements taken during a diagnostic session; (b) current ranges assigned to different currents; (c) a diagnosis or tentative diagnosis that specifies one or more medical conditions that are indicated by the electrical current measurements taken during the diagnostic session; (d) a confidence level or probability associated with each diagnosis or tentative diagnosis; (e) one or more recommendations for action to be taken (e.g., a recommendation to check with a physician for further testing or for confirmation or treatment of a medical condition); (f) additional information about the diagnostic process and the current measurements; (g) results of previous diagnostic sessions; and (h) a comparison of a current diagnosis (or diagnoses) with a past diagnosis (or diagnoses).

The GUI or UI may include a chat box. The chat box may enable a patient to provide additional information about symptoms and to ask questions. In some cases, the chatbox will enable a patient to select from a list of symptoms, and also enable the patient to input information about symptoms that are not listed. The chat box may also enable a health care worker to ask additional questions and to receive answers from the patient.

One or more computers may employ a chatbot in a UI, in order to gather input from and provide information to a patient, health care worker or other user. In some cases, at least some of the information that is provided to a patient, health care worker or other user is sent via one or more emails or other social media messages. The information that is provided by chatbot, email or other social media message may comprise any or all of the information described above in this “User Interface” section.

In some cases, an audiovisual UI guides a user (e.g., patient or health care worker) to take the electrical current measurements under conditions that are suitable for accurate readings. Put differently, the audiovisual UI may provide real-time feedback regarding whether the electrodes are properly positioned and pressed firmly enough against the skin.

When an electrode is pressed against a patient’s skin, the measured electrical current may increase as the pressure exerted against the skin increases, until the measured electrical current reaches a plateau. In some cases: (a) the current sensor detects when the electric current is increasing and when the current plateaus; (b) the only measurements of electric current that are used for diagnostic purposes occur after the measured current has increased and reached a plateau, and (c) measurements of electric current that are taken before the current reaches a plateau are disregarded for diagnostic purposes. Alternatively or in addition, in some cases: (a) one or more pressure sensors measure pressure exerted against an electrode; (b) the only measurements of electric current that are used for diagnostic purposes occur when the pressure exerted on the electrode exceeds a threshold value; and (c) measurements of electric current that are taken when the pressure exerted on the electrode is less than or equal to the threshold are disregarded for diagnostic purposes. In some cases, an electrode has multiple pads, and electric current measurements are disregarded for diagnostic purposes unless the pressure exerted against a threshold number of the pads exceeds a threshold pressure. In some cases, pressure is measured for both the ground electrode and probe electrode, and electric current measurements are disregarded for diagnostic purposes unless pressure exerted against each electrode exceeds the threshold pressure for that electrode. In some cases, the same pressure threshold is used for both the ground and probe electrodes and for all of the pads of a ground electrode. Alternatively, different pressure thresholds may be employed for different electrodes and/or for different pads of an electrode.

The audiovisual UI may emit a sound (e.g., a beep or tone) when the probe and ground electrodes are held correctly. The UI may include a sonic guide that changes pitch or tone depending on the amount of pressure applied to electrode(s) or depending on the amount of current being detected. The UI may also include a visual indicator that shows whether each electrode is in proper contact with the patient’s skin. For instance, the graphic display may highlight which electrode is not properly contacting the patient’s skin, by changing the color or shape of an electrode icon on the screen.

Thus, the UI may enable self-calibrated measurements of cross-body electrical currents on patients of varying measurement location physiologies (e.g., sizes, skin thickness, exact probe placement, variations in galvanic skin response and transient surface currents effects) and may be used to rapidly screen for optimal measurement protocols. In some implementations, the measurement protocol includes both: (a) real- time determination of quality of signal; and (b) real-time feedback to a user via the audiovisual UI.

Customization

In some cases, the machine learning algorithm is trained on a dataset for a general population.

In other cases, the machine learning algorithm is trained to predict medical conditions in a way that is customized for one or more features of a patient, such as the patient’s age, sex, race, weight, habits (e.g., smoker vs non-smoker), personal medical history and/or family medical history. For instance, the training dataset for the machine learning algorithm may be labeled with not only medical conditions but also with one or more these features (e.g., patient’s age, sex, race, weight, habits, personal medical history and/or family medical history).

Likewise, if a lookup table is employed instead of a machine learning algorithm, then multiple lookup tables may be used, each customized for a different combination of these features. As a non-limiting example, there may be a first lookup table for males over age 59, a second lookup table for women over age 59, a third lookup table for males age 31-59, and so on.

In some implementations of this invention, a machine learning model is personalized for a particular patient. For instance, a machine learning algorithm may be initially trained on data for a general population or for a subset of a general population. Then the machine learning algorithm may be further trained for a particular patient, based on data gathered in the course of performing diagnoses of the particular patient. For instance, if the machine learning algorithm predicts medical condition A for a patient but the patient actually has medical condition B, then this information may be used as part of an additional training dataset to train the machine learning algorithm to make personalized predictions for the patient.

Likewise, if a lookup table is employed instead of a machine learning algorithm, then the lookup table may be personalized, based on data gathered in the course of making diagnoses of the particular patient.

Adaptive Prediction

In some cases, the electrical current measurements for a patient are supplemented with information about contextual features. For instance, the contextual information may include sensor readings that are taken by one or more sensors which are worn by, or located near to, the patient. These other sensors may measure one or more physiological states of the patient (e.g., heart rate, respiration rate, body temperature) and/or one or more states of the patient’s environment (e.g., temperature, humidity, ambient light). These other sensors may wirelessly transmit their readings to a receiver in the diagnostic system. In some cases, the contextual information also includes text or audio input from a patient or health care worker regarding the patient’s state (e.g., happy, worried) and/or the patient’s environment (e.g., at work).

In some cases, the machine learning algorithm is trained to adapt its prediction in real time based on data about the patient’s context. For instance, the training dataset for the machine learning algorithm may be labeled with not only medical conditions but also with one or more contextual features (such as one or more physiological states, mental states, and environmental features). Likewise, if a lookup table is employed, different versions of the lookup table may be employed, depending on the patient’s context.

After a machine learning model is initially trained, it may adaptively learn based on the patient’s context when electric current measurements are taken. Data regarding both the electrical currents and the context may be gathered while making diagnoses and may later be employed as an additional training dataset, in order to further train the model to predict medical conditions in a manner that depends in part on context.

Machine Learning Example

The following 23 paragraphs describe an example (the “ML Example”) of a diagnostic system that employs a trained machine learning model. The ML Example is a non-limiting example of this invention.

In this ML Example, a diagnostic system captures, organizes and analyzes measurements of electrical currents. The system employs machine learning and a database (knowledge library). The system may be used by less experienced practitioners to quickly and accurately diagnose their patients.

In this ML Example: (a) electrical current (or conductivity of the skin) is measured at each meridian point; and (b) excessive and deficient energies are plotted on a chart and identified. Treatment may consist of stimulating specific acupuncture points to either “tonify” a deficient meridian, or “sedate” an excessive meridian.

In this ML Example, patients have different electrical conductivity potentials. Thus, in this ML Example, current measurements are not absolute, but rather may be taken relative to all other measurements on the same patient. Thus, deviation from an average measurement may be more important than the actual measurement itself. Deviations (which may be used for diagnosis) may be determined by analyzing measurements within a broader context (e.g., plotting the current measurements on a chart and looking for outliers from the mean). For instance, an analysis may be designed to encompass the majority of measurements, in an area we sometimes call a “physiological corridor.” Measurements outside the corridor may be deemed abnormal, and treatment applied to restore balance to abnormal meridians.

In this ML example, screening may be employed to identify the possible presence of an as-yet-undiagnosed disease in an individual patient (e.g., without signs or symptoms). This may include individuals with pre-symptomatic or unrecognized symptomatic disease.

In this ML Example, electrical measurements along acupuncture meridian lines may be used to examine and identify an individual’s specific areas of weakness and strength in order determine a condition, disease or illness. The electrical conductance of the primary meridian lines may be measured at various points on the patient’s wrists and ankles. Both excessive and deficient electrical conductance levels outside the patient’s normal range may be correlated to classify the condition of the patient.

In this ML Example, a differential diagnostic process may distinguish a particular condition from others that present similar symptoms. A differential diagnosis may include the following steps: (a) gather information about the patient to be diagnosed and create a symptoms list; (b) list possible causes (candidate conditions) for the symptoms; (c) prioritize the list by placing the most urgently dangerous condition at the top of the list; (d) work down the list to rule out possible causes; and (e) remove diagnoses from the list by observing and applying tests that produce different results.

In this ML Example, meridian point assessment of the patient’s condition may be used to assemble and support possible candidate conditions and also potentially rule out other possible causes from consideration.

In this ML Example, the diagnostic system may be applied to assess the mental health status of a patient.

In this ML Example, the mental health or psychiatric condition of the patient’s mind may have adverse effects on the patient’s body. For example, anxiety or depression (rather than an infection or physical abnormality in the digestive tract) may be the root cause of dyspepsia. The conductivity measurements may also provide data regarding psychological aspects of an individual, not just the physical.

In this ML Example, the diagnostic system may be a decision/support system that: (a) links observations with a database of knowledge; and (b) helps to analyze the current state of a patient and to reach a diagnostic conclusion.

FIG. 12 is a flowchart for a diagnostic method employed in the ML Example. The method shown in FIG. 12 includes at least the following steps: capture 1210, machine learning 1220, and prediction 1230.

In the ML Example, the capture step may comprise: (a) taking patient observations in the form of recorded meridian points; and (b) producing a database of associated labeled outcomes for a selected diagnosis (e.g., where the outcome labels are Positive, Negative and Rule Out). The information acquired during capture may be used to create the knowledge library database. Each patient record may consist of 24 meridian points, 12 from the left and right hands and 12 from the left and right feet.

In the ML Example, the machine learning step may include generating a set of random forests through supervised training, in such a way that: (a) one random forest is created for each potential diagnostic entity; and (b) the collection of random forests constitutes the knowledge library database.

In the ML Example, each random forest may be employed as an ensemble of knowledge for a given diagnostic conclusion. The conclusion may be either positive, negative or needs further testing to rule out. Each trained ensemble may represent a single hypothesis.

In the ML Example, any type of machine learning model may be employed, including: (a) an artificial neural network; (b) decision tree; (c) random forests; or (d) support vector machine.

In the ML Example, the machine learning model is trained by supervised learning. For example, the experience (and/or separate diagnosis) data entered by a doctor may be organized into the content of the models. Thus, a doctor may supervise the learning of the machine learning model.

In the version of the ML Example that is shown in FIG. 12 , a doctor may enter a set of outcomes (positive, negative and/or rule out) for a specific set of patient meridian points as it relates to a specific diagnosis. Each experience is recorded in the database. In the supervised learning, the patient meridian points may be features and the outcomes may be labels. During supervised learning, a decision tree may split data into smaller data groups based on the features of the data until a small enough set of data identifies to one label. After the decision tree is trained, it may take as an input a feature set (meridian points) and may output one label (positive, negative or rule out).

In the ML Example, rather than rely on only one decision tree, a random forest is created that consists of a number of competing decision trees, where each tree is trained in a slightly different way. Then each tree in the forest may determine an answer on its own and the forest may be surveyed for the best agreed upon answer.

In the ML Example, the supervised learning mode may output an accurate predicted label (outcome). Meridian points recorded from a new patient may be entered into the diagnostic system and a patient diagnosis may be displayed on a practitioner’s monitor.

In the ML Example, the machine learning algorithm may create a random forest for each diagnostic candidate and each forest may consist of hundreds of trees. As a non-limiting example, if there are a hundred diagnostic candidates, then there may be tens of thousands of decision trees that are making decisions (e.g., correlating meridian point data to possible outcomes).

In the ML Example, an inference engine may query each random forest for its outcome decision. The inference engine may assess the reliability of each decision, rank them and present them. The inference engine may also present supporting justification for the final set of outcomes. The inference engine may also record feedback from the practitioner to determine the validity of the final outcome. The information may be recorded and ultimately fed back into the machine learning algorithm to enhance system performance and accuracy.

In the ML Example, data may be stored in a relational database. For instance, data may be stored in the relational database shown in FIG. 13 . This relational database may include data regarding, among other things, patients 1300, conditions 1310, positive outcomes 1320, negative outcomes 1330, and rule-outs 1340.

In the ML Example, each meridian record may be analyzed against the various diagnostic candidates and an outcome of positive, negative or rule out may be determined. A graphical user interface may display diagnostic results and may also display a justification for the diagnosis.

The ML Example described in the preceding 23 paragraphs is a non- limiting example of this invention. This invention may be implemented in many other ways.

Practical Applications

This invention has many practical applications. For instance, in some cases, the diagnostic system may be employed to diagnose or tentatively diagnose a medical condition. The diagnostic system may also be employed to screen for medical conditions, and to determine when further testing is needed in order to determine whether a particular medical condition is present. In some implementations, the diagnostic system is employed to quickly distinguish between a viral infection and a bacterial infection. Also, in some cases, the diagnostic system may be employed to rapidly screen for: (a) optimal dosing levels for medicine; (b) effects of (physical or psycho-) therapy or exercise; (c) effects of diet or other therapeutic or preventative or wellness-focused supplements; and (d) effects of pharmaceuticals and/or other therapeutic and diagnostic interventions.

Computers

In illustrative implementations of this invention, one or more computers (e.g., servers, network hosts, client computers, integrated circuits, microcontrollers, controllers, microprocessors, field-programmable-gate arrays, personal computers, digital computers, driver circuits, or analog computers) are programmed or specially adapted to perform one or more of the following tasks: (1) to control the operation of, or interface with, hardware components of a current sensor, power source, or signal generator; (2) to calibrate, filter and/or average current measurements; (3) to calculate current ranges and to assign currents to current ranges; (4) to determine an electrical current state that consists of a current range for a specific current or of current ranges for respective currents; (5) to access a lookup table to determine that one or more medical conditions are indicated by the electrical current state; (6) to train a machine learning model; (7) to employ a trained machine learning model to predict, based on measured cross-body electrical currents, that one or more medical conditions are present; (8) to output a diagnosis or tentative diagnosis; (9) to output a rule-out recommendation to perform further medical testing to evaluate whether a medical condition is actually present; (10) to output a probability or confidence level for each medical condition that is diagnosed or tentatively diagnosed; (11) to control input/output devices in such as to present a UI that provides real-time feedback regarding whether electrodes are being used properly and that provides other information including current readings and diagnoses; (12) to receive data from, control, or interface with one or more sensors, including one or more pressure sensors; (13) to perform any other calculation, computation, program, algorithm, or computer function described or implied herein; (14) to receive signals indicative of human input; (15) to output signals for controlling transducers for outputting information in human perceivable format; (16) to process data, to perform computations, and to execute any algorithm or software; and (17) to control the read or write of data to and from memory devices (tasks 1-17 of this sentence being referred to herein as the “Computer Tasks”). The one or more computers (e.g. 105, 121 or a computer in smartphone 450) may, in some cases, communicate with each other or with other devices: (a) wirelessly, (b) by wired connection, (c) by fiber-optic link, or (d) by a combination of wired, wireless or fiber optic links.

In exemplary implementations, one or more computers are programmed to perform any and all calculations, computations, programs, algorithms, computer functions and computer tasks described or implied herein. For example, in some cases:

(a) a machine-accessible medium has instructions encoded thereon that specify steps in a software program; and (b) the computer accesses the instructions encoded on the machine-accessible medium, in order to determine steps to execute in the program. In exemplary implementations, the machine-accessible medium may comprise a tangible non-transitory medium. In some cases, the machine-accessible medium comprises (a) a memory unit or (b) an auxiliary memory storage device. For example, in some cases, a control unit in a computer fetches the instructions from memory.

In illustrative implementations, one or more computers execute programs according to instructions encoded in one or more tangible, non-transitory computer-readable media. For example, in some cases, these instructions comprise instructions for a computer to perform any calculation, computation, program, algorithm, or computer function described or implied herein. For instance, in some cases, instructions encoded in a tangible, non-transitory, computer-accessible medium comprise instructions for a computer to perform the Computer Tasks.

Computer Readable Media

In some implementations, this invention comprises one or more computers that are programmed to perform one or more of the Computer Tasks.

In some implementations, this invention comprises one or more tangible, machine readable media, with instructions encoded thereon for one or more computers to perform one or more of the Computer Tasks. In some implementations, these one or more media are not transitory waves and are not transitory signals.

In some implementations, this invention comprises participating in a download of software, where the software comprises instructions for one or more computers to perform one or more of the Computer Tasks. For instance, the participating may comprise (a) a computer providing the software during the download, or (b) a computer receiving the software during the download.

Network Communication

In illustrative implementations of this invention, one or more devices (e.g., 105, 450) are configured for wireless or wired communication with other devices in a network.

For example, in some cases, one or more of these devices include a wireless module for wireless communication with other devices in a network. Each wireless module may include (a) one or more antennas, (b) one or more wireless transceivers, transmitters or receivers, and (c) signal processing circuitry. Each wireless module may receive and transmit data in accordance with one or more wireless standards.

In some cases, one or more of the following hardware components are used for network communication: a computer bus, a computer port, network connection, network interface device, host adapter, wireless module, wireless card, signal processor, modem, router, cables and wiring.

In some cases, one or more computers (e.g., 105 or a computer in smartphone 450) are programmed for communication over a network. For example, in some cases, one or more computers are programmed for network communication: (a) in accordance with the Internet Protocol Suite, or (b) in accordance with any other industry standard for communication, including any USB standard, ethernet standard (e.g., IEEE 802.3), token ring standard (e.g., IEEE 802.5), or wireless communication standard, including IEEE 802.11 (Wi-Fi®), IEEE 802.15 (Bluetooth®/Zigbee®), IEEE 802.16, IEEE 802.20, GSM (global system for mobile communications), UMTS (universal mobile telecommunication system), CDMA (code division multiple access, including IS-95, IS-2000, and WCDMA), LTE (long term evolution), or 5G (e.g., ITU IMT-2020).

Definitions

The terms “a” and “an”, when modifying a noun, do not imply that only one of the noun exists. For example, a statement that “an apple is hanging from a branch”: (i) does not imply that only one apple is hanging from the branch; (ii) is true if one apple is hanging from the branch; and (iii) is true if multiple apples are hanging from the branch.

“Associate” is defined above.

To compute “based on” specified data means to perform a computation that takes the specified data as an input.

To say that a current flows “between” A and B does not create any implication regarding direction of flow (i.e., from A to B, or from B to A).

“BL current” is defined above.

Non-limiting examples of a “camera” include: (a) a digital camera; (b) a digital grayscale camera; (c) a digital color camera; and (d) a video camera.

The term “comprise” (and grammatical variations thereof) shall be construed as if followed by “without limitation”. If A comprises B, then A includes B and may include other things.

A digital computer is a non-limiting example of a “computer”. An analog computer is a non-limiting example of a “computer”. A computer that performs both analog and digital computations is a non-limiting example of a “computer”. However, a human is not a “computer”, as that term is used herein.

“Computer Tasks” is defined above.

“Defined Term” means a term or phrase that is set forth in quotation marks in this Definitions section.

For an event to occur “during” a time period, it is not necessary that the event occur throughout the entire time period. For example, an event that occurs during only a portion of a given time period occurs “during” the given time period.

The term “e.g.” means for example.

The fact that an “example” or multiple examples of something are given does not imply that they are the only instances of that thing. An example (or a group of examples) is merely a non-exhaustive and non-limiting illustration.

Chronic fatigue is a non-limiting example of “fatigue”.

The terms “Class A Condition” through “Class N Condition” are defined above. Also, “Class P Condition” and “Class Q Condition” are defined above.

“Diagnostic session” means a period of time.

Unless the context clearly indicates otherwise: (1) a phrase that includes “a first” thing and “a second” thing does not imply an order of the two things (or that there are only two of the things); and (2) such a phrase is simply a way of identifying the two things, so that they each may be referred to later with specificity (e.g., by referring to “the first” thing and “the second” thing later). For example, if a device has a first socket and a second socket, then, unless the context clearly indicates otherwise, the device may have two or more sockets, and the first socket may occur in any spatial order relative to the second socket. A phrase that includes a “third” thing, a “fourth” thing and so on shall be construed in like manner.

As used herein, “food-related sinus allergy” means a sinus allergy that is caused (or exacerbated) at least in part by one or substances (e.g., allergens) in ingested food.

“Forearm” is defined above.

“For instance” means for example.

To say a “given” X is simply a way of identifying the X, such that the X may be referred to later with specificity. To say a “given” X does not create any implication regarding X. For example, to say a “given” X does not create any implication that X is a gift, assumption, or known fact.

A migraine is a non-limiting example of a “headache”.

“Herein” means in this document, including text, specification, claims, abstract, and drawings.

As used herein: (1) “implementation” means an implementation of this invention; (2) “embodiment” means an embodiment of this invention; (3) “case” means an implementation of this invention; and (4) “use scenario” means a use scenario of this invention.

To say that a current is “in” a patient means that the current flows through at least a portion of the body of the patient.

The term “include” (and grammatical variations thereof) shall be construed as if followed by “without limitation”.

“GB current” is defined above.

“HT current” is defined above.

“KI current” is defined above.

“Left side” is defined above.

“Leg” is defined above.

“LI current” is defined above.

As used herein, “lower back” means the portion of the back that is inferior to the transpyloric plane.

“LR current” is defined above.

“LU current” is defined above.

A physiological condition is a non-limiting example of a “medical condition”, as that term is used herein.

“Meridian” means acupuncture meridian.

The term “mobile computing device” or “MCD” means a device that includes a computer, a camera, a display screen and a wireless transceiver. Non-limiting examples of an MCD include a smartphone, cell phone, mobile phone, tablet computer, laptop computer and notebook computer.

Unless the context clearly indicates otherwise, “or” means and/or. For example, A or B is true if A is true, or B is true, or both A and B are true. Also, for example, a calculation of A or B means a calculation of A, or a calculation of B, or a calculation of A and B.

“PC current” is defined above.

As used herein, “poor glycemic control” means: (a) blood glucose levels that are persistently greater than 200 mg/dl; together with (b) glycated hemoglobin levels in the blood that are persistently greater than 9%.

“Prototype Current” is defined above.

“Prototype Current Ranges” is defined above.

“Prototype Electrical Current State” is defined above.

“Prototype Measurement Locations” is defined above.

“Prototype Medical Condition” is defined above.

“Right side” is defined above.

As used herein, the term “set” does not include a group with no elements.

“SI current” is defined above.

An electrode touching or being pressed against a conductive gel (or other conductive material) that is on a region of skin of a patient is a non-limiting example of the electrode “touching” or being “pressed against” the region of skin, as those terms are used herein.

Unless the context clearly indicates otherwise, “some” means one or more.

“SP current” is defined above.

“ST current” is defined above.

As used herein, a “subset” of a set consists of less than all of the elements of the set.

The term “such as” means for example.

“TH current” is defined above.

To say that a current flows “through” a body means that the current flows through at least a portion of the body.

To say that a machine-readable medium is “transitory” means that the medium is a transitory signal, such as an electromagnetic wave.

As used herein, “upper back” means the portion of the back that is superior to the transpyloric plane.

In the clause “HT current is above average or below average on left side”, the phrase “on left side” modifies both “above average” and “below average”. Likewise, other clauses with the same grammatical structure shall be construed in the same way. For instance, in the clause “LI current is average or below average on right side”, the phrase “on right side” modifies both “average” and “below average”.

“A xor B” means A or B but not A and B. Put differently, the term “xor” signifies an exclusive or.

Except to the extent that the context clearly requires otherwise, if steps in a method are described herein, then the method includes variations in which: (1) steps in the method occur in any order or sequence, including any order or sequence different than that described herein; (2) any step or steps in the method occur more than once; (3) any two steps occur the same number of times or a different number of times during the method; (4) one or more steps in the method are done in parallel or serially; (5) any step in the method is performed iteratively; (6) a given step in the method is applied to the same thing each time that the given step occurs or is applied to a different thing each time that the given step occurs; (7) one or more steps occur simultaneously; or (8) the method includes other steps, in addition to the steps described herein.

Headings are included herein merely to facilitate a reader’s navigation of this document. A heading for a section does not affect the meaning or scope of that section.

This Definitions section shall, in all cases, control over and override any other definition of the Defined Terms. The Applicant or Applicants are acting as his, her, its or their own lexicographer with respect to the Defined Terms. For example, the definitions of Defined Terms set forth in this Definitions section override common usage and any external dictionary. If a given term is explicitly or implicitly defined in this document, then that definition shall be controlling, and shall override any definition of the given term arising from any source (e.g., a dictionary or common usage) that is external to this document. If this document provides clarification regarding the meaning of a particular term, then that clarification shall, to the extent applicable, override any definition of the given term arising from any source (e.g., a dictionary or common usage) that is external to this document. Unless the context clearly indicates otherwise, any definition or clarification herein of a term or phrase applies to any grammatical variation of the term or phrase, taking into account the difference in grammatical form. For example, the grammatical variations include noun, verb, participle, adjective, and possessive forms, and different declensions, and different tenses.

Table 2, below, provides exemplary data measurements obtained from 133 people, and associated diagnoses, rule-outs, and symptoms, if any. The measurements were taken as herein described at the respective lung, pericardium, heart, small intestine, triple heater, large intestine, spleen, liver, kidney, bladder, gall bladder, and stomach acupuncture meridian locations. The measurements are provided in units of microamps (µA).

TABLE 2 Patient 1 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 74 66 78 62 144 154 76 54 66 76 46 92 Right: LU PC HT SI TH LI SP LR KI BL GB ST 72 46 50 170 186 176 48 58 46 68 36 108 Diagnosis: Group #1: nerve damage Group #1: nerve damage — upper and lower back Group #1: microcirculatory problem Group #1: microcirculatory-orthostatic hypotension Group #2: low protein intake Group #2: increased metabolism Group #2: weight loss Group #3: stress Rule Outs: Group #1: coronary artery disease Symptoms: Group #1: chronic fatigue Patient 2 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 112 114 102 26 64 78 184 82 116 132 86 128 Right: LU PC HT SI TH LI SP LR KI BL GB ST 152 126 118 120 126 96 116 98 134 132 72 112 Diagnosis: Group #1: lower and upper back irritation Group #2: low protein intake Symptoms: Group #1: lower back pain Patient 3 Measurments: Left: LU PC HT SI TH LI SP LR KI BL GB ST 82 62 66 70 110 64 86 42 38 64 46 72 Right: LU PC HT SI TH LI SP LR KI BL GB ST 52 62 50 76 78 66 82 80 46 68 42 66 Diagnosis: Group #1: headaches Group #1: nerve damage — neck Group #2: hyperthyroid Rule Outs: Group #2: hyperthyroid Patient 4 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 24 42 26 14 20 20 92 56 48 108 46 42 Right: LU PC HT SI TH LI SP LR KI BL GB ST 68 36 22 28 26 16 148 118 86 106 78 4 Diagnosis: Group #1: lower and upper back irritation Group #2: gastroparesis Group #3: fatigue Group #3: low blood pressure Group #4: depression Group #5: high carbohydrate intake Group #6: sinus congestion Patient 5 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 24 42 26 14 20 20 92 56 48 108 46 42 Right: LU PC HT SI TH LI SP LR KI BL GB ST 68 36 22 28 26 16 148 118 86 106 78 0 Diagnosis: Group #1: gastroparesis Group #2: nerve related slow peristaltic activity Group #3: low blood pressure Group #4: lower and upper back irritation Group #5: depression Group #5: chronic fatigue Group #6: sinus congestion Rule Outs: Group #6: sinus infection Group #7: depression Patient 6 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 16 8 24 86 128 124 138 192 102 122 46 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 74 36 18 108 108 100 142 200 90 130 200 200 Diagnosis: Group #1: low blood pressure Group #2: sinus congestion Group #2: asthma Group #3: reflux Group #4: helicobacter infection of stomach Group #5: shoulder pain Group #6: lower and upper back irritation Patient 7 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 186 182 140 102 170 188 172 44 136 158 174 58 Right: LU PC HT SI TH LI SP LR KI BL GB ST 150 184 174 200 200 182 182 40 122 130 132 40 Diagnosis: Group #1: gastroparesis Group #2: low protein intake Group #2: low stomach function Group #2: weight loss Group #3: stress Group #4: neck damage Group #5: microcirculatory problem Group #6: post nasal drip Patient 8 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 172 196 188 46 68 60 102 78 130 130 94 108 Right: LU PC HT SI TH LI SP LR KI BL GB ST 158 108 88 80 58 46 94 108 116 132 90 126 Diagnosis: Group #1: lower and upper back irritation Group #2: stress Group #3: post nasal drip Group #4: high carbohydrate intake Rule Outs: Group #3: lung cancer Patient 9 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 134 112 198 112 132 144 134 160 120 70 128 196 Right: LU PC HT SI TH LI SP LR KI BL GB ST 120 102 124 170 176 174 108 150 144 82 134 184 Diagnosis: Group #1: lower back injury Group #1: neck injury Group #2: microcirculatory problem Group #2: fatigue Group #2: sleep disturbance Group #3: stomach irritation Patient 10 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 114 92 70 66 32 70 142 76 80 102 80 78 Right: LU PC HT SI TH LI SP LR KI BL GB ST 96 94 60 58 52 56 112 80 52 96 64 90 Diagnosis: Group #1: lower and upper back irritation Group #1: nerve damage — neck Group #2: post nasal drip Group #3: hyperthyroid Patient 11 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 148 124 152 174 176 180 108 174 154 52 152 174 Right: LU PC HT SI TH LI SP LR KI BL GB ST 182 146 136 168 172 190 86 134 150 56 138 142 Diagnosis: Group #1: nerve damage — lower back Group #1: bladder disfunction Group #1: microcirculatory problem Group #1: fatigue Group #1: sleep disturbance Group #2: increased metabolism Group #3: anemia or low blood count Patient 12 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 138 88 108 128 120 116 148 148 88 120 80 100 Right: LU PC HT SI TH LI SP LR KI BL GB ST 118 86 90 146 118 124 172 128 80 128 194 182 Diagnosis: Group #1: nerve damage — neck Group #1: microcirculatory-orthostatic hypotension Group #1: microcirculatory problem Group #1: sleep disturbance Group #1: fatigue Group #2: reflux Group #2: shoulder pain Patient 13 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 190 190 200 142 148 162 154 186 158 142 124 150 Right: LU PC HT SI TH LI SP LR KI BL GB ST 182 152 122 194 182 166 178 144 176 124 118 166 Diagnosis: Group #1: back injury Group #2: depression Patient 14 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 100 38 20 24 30 56 82 140 34 44 70 148 Right: LU PC HT SI TH LI SP LR KI BL GB ST 90 18 20 20 50 44 60 96 54 36 158 134 Diagnosis: Group #1: nerve damage — lower back Group #1: nerve damage — neck Group #2: microcirculatory-orthostatic hypotension Patient 15 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 34 26 46 10 80 18 38 22 36 22 10 10 Right: LU PC HT SI TH LI SP LR KI BL GB ST 24 16 14 28 44 36 46 16 42 34 8 16 Diagnosis: Group #1: fatigue Patient 16 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 154 128 162 128 104 124 158 88 118 130 36 128 Right: LU PC HT SI TH LI SP LR KI BL GB ST 160 126 128 118 112 124 188 190 118 130 166 132 Diagnosis: Group #1: post nasal drip Group #2: depression Group #3: sinus infection Group #4: hypothyroid disease Patient 17 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 88 82 88 46 54 68 110 126 26 18 24 30 Right: LU PC HT SI TH LI SP LR KI BL GB ST 150 60 88 72 88 92 68 110 36 18 36 104 Diagnosis: Group #1: sinus allergy Group #2: depression Group #3: nerve damage — lower back Group #3: nerve damage — neck Group #4: bladder dysfunction Patient 18 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 86 98 70 118 142 154 18 16 8 10 30 34 Right: LU PC HT SI TH LI SP LR KI BL GB ST 80 54 48 102 108 154 28 54 20 16 24 40 Diagnosis: Group #1: nerve damage — upper and lower back Group #2: migraines Group #3: weight loss Group #4: MS Group #4: bladder dysfunction Patient 19 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 28 36 36 30 50 48 34 170 44 28 140 130 Right: LU PC HT SI TH LI SP LR KI BL GB ST 48 52 38 48 34 32 56 132 28 26 104 142 Diagnosis: Group #1: microcirculatory-orthostatic hypotension Group #1: nerve damage — upper and lower back Group #1: bladder dysfunction Group #2: food related sinus allergy Group #3: spicy or high fat injestion Group #3: high carbohydrate intake Group #3: high fat intake Group #4: high carbohydrate intake Group #4: high fat intake Patient 20 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 174 136 104 150 148 170 136 106 152 128 110 138 Right: LU PC HT SI TH LI SP LR KI BL GB ST 162 130 122 138 152 130 122 94 112 120 122 138 Diagnosis: Group #1: low protein intake Group #2: neck injury Group #2: nerve damage — neck Group #2: microcirculatory problem Group #2: fatigue Group #3: post nasal drip Group #3: lung cancer Patient 21 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 182 192 192 70 116 30 110 58 64 96 60 56 Right: LU PC HT SI TH LI SP LR KI BL GB ST 184 170 184 94 118 120 98 84 70 114 92 58 Diagnosis: Group #1: high stress stress Group #1: anxiety Group #1: hypertension Group #2: nerve damage — neck Group #2: kidney failure Group #3: large bowel issues Group #3: low stomach function Group #3: low bowel movements Patient 22 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 120 76 90 94 58 52 66 68 98 78 86 138 Right: LU PC HT SI TH LI SP LR KI BL GB ST 132 86 84 126 100 126 94 152 74 48 122 148 Diagnosis: Group #1: microcirculatory problem Group #1: neck damage Group #1: lower back injury Group #1: fatigue Group #1: bladder dysfunction Group #2: reflux Group #3: sinus irritation Group #3: sinus allergy Patient 23 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 142 88 96 134 130 116 82 126 96 58 20 84 Right: LU PC HT SI TH LI SP LR KI BL GB ST 130 134 124 130 100 112 134 130 104 58 86 104 Diagnosis: Group #1: nerve damage — lower back Group #1: fatigue Group #1: bladder dysfunction Group #1: lower back pain Group #2: sinus allergy Group #2: sinus irritation Group #3: depression Patient 24 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 78 40 38 20 34 24 88 50 78 116 82 106 Right: LU PC HT SI TH LI SP LR KI BL GB ST 60 48 50 22 32 26 102 94 82 116 16 0 Diagnosis: Group #1: upper and lower back irritation Group #1: bladder dysfunction Group #2: fatigue Group #2: coronary artery disease Group #2: low blood pressure Group #3: male Group #3: possible prostate enlargement Group #3: prostate cancer Group #3: lower back injury Patient 25 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 116 86 86 110 58 90 116 42 104 146 72 74 Right: LU PC HT SI TH LI SP LR KI BL GB ST 142 118 106 66 76 68 96 34 68 122 84 64 Diagnosis: Group #1: lower back injury Group #1: back injury Group #1: prostate cancer Group #2: nerve damage — neck Group #2: bladder dysfunction Group #3: post nasal drip Group #3: lung cancer Group #4: low protein intake Group #5: hypothyroid disease Patient 26 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 88 52 88 60 30 52 76 92 64 98 38 94 Right: LU PC HT SI TH LI SP LR KI BL GB ST 80 52 92 54 40 72 114 118 70 64 30 80 Diagnosis: Group #1: depression Group #1: fatigue Group #2: lower back injury Group #2: lower back pain Group #3: depression Patient 27 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 48 76 92 58 56 34 60 134 90 76 46 102 Right: LU PC HT SI TH LI SP LR KI BL GB ST 90 44 74 62 66 24 46 90 60 58 86 88 Diagnosis: Group #1: food related sinus allergy Group #2: microcirculatory problem Group #2: depression Group #3: large bowel issues Patient 28 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 86 74 32 88 62 64 46 58 84 104 18 72 Right: LU PC HT SI TH LI SP LR KI BL GB ST 126 58 58 62 26 62 80 86 80 104 90 42 Diagnosis: Group #1: coronary artery disease Group #2: post nasal drip Group #3: lower back injury Group #3: lower back pain Group #4: male Group #4: prostate cancer Patient 29 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 68 66 78 38 60 46 104 74 68 118 122 68 Right: LU PC HT SI TH LI SP LR KI BL GB ST 106 56 78 54 106 84 144 80 70 86 160 140 Diagnosis: Group #1: lower and upper back irritation Group #1: shoulder pain Group #1: bladder dysfunction Group #2: reflux Group #2: shoulder pain Group #2: stomach irritation Patient 30 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 150 152 96 8 34 26 136 140 134 132 52 140 Right: LU PC HT SI TH LI SP LR KI BL GB ST 166 152 96 34 34 22 126 68 128 128 96 122 Diagnosis: Group #1: high carbohydrate intake Group #1: low protein intake Group #2: lower and upper back irritation Group #2: neck irritation Group #2: bladder dysfunction Group #3: post nasal drip Group #3: esophageal obstruction Group #3: food related sinus allergy Patient 31 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 96 78 106 66 18 34 70 54 62 84 58 132 Right: LU PC HT SI TH LI SP LR KI BL GB ST 84 46 62 84 42 48 82 132 52 66 64 114 Diagnosis: Group #1: depression Group #2: fatigue Group #3: post nasal drip Group #3: sinus allergy Group #4: stomach irritation Group #5: hyperthyroid Group #6: migraines Patient 32 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 158 164 170 66 62 98 174 78 158 176 98 118 Right: LU PC HT SI TH LI SP LR KI BL GB ST 180 170 160 66 84 64 188 128 176 164 106 102 Diagnosis: Group #1: hypertension Group #1: stress Group #2: high carbohydrate intake Group #2: low protein intake Group #3: lower and upper back irritation Group #3: bladder infection Group #3: bladder dysfunction Patient 33 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 80 74 46 56 50 80 44 68 84 72 34 78 Right: LU PC HT SI TH LI SP LR KI BL GB ST 116 84 68 78 64 62 74 34 40 98 76 60 Diagnosis: Group #1: nerve damage — neck Group #3: lower and upper back irritation Group #4: post nasal drip Patient 34 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 118 114 96 152 166 112 40 12 54 68 28 28 Right: LU PC HT SI TH LI SP LR KI BL GB ST 146 74 72 198 180 190 82 8 58 62 72 92 Diagnosis: Group #1: weight loss Group #1: low protein intake Group #2: post nasal drip Group #3: nerve damage — neck Group #3: nerve damage — lower back Group #3: bladder dysfunction Patient 35 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 32 6 12 34 58 106 34 102 52 40 78 92 Right: LU PC HT SI TH LI SP LR KI BL GB ST 30 8 36 34 32 52 76 76 62 20 116 160 Diagnosis: Group #1: hypotension Group #1: microcirculatory-orthostatic hypotension Group #1: blood pressure medications overdose Group #2: nerve damage — lower back Group #2: bladder dysfunction Group #3: sinus congestion Group #4: large bowel issues Patient 36 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 200 200 172 96 136 106 126 164 138 120 116 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 200 200 190 114 114 124 172 200 108 128 198 200 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: bladder dysfunction Group #2: reflux Group #3: post nasal drip Group #4: depression Group #5: trouble swallowing Patient 37 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 138 144 128 144 108 188 156 192 122 128 170 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 160 134 86 166 88 96 152 200 92 136 158 200 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: bladder dysfunction Group #2: coronary artery disease Group #3: food related stomach irritation (spicy, cheese) Group #4: hypothyroid disease Patient 38 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 56 56 50 44 30 36 86 26 70 90 26 26 Right: LU PC HT SI TH LI SP LR KI BL GB ST 88 70 76 30 22 22 116 40 64 122 24 18 Diagnosis: Group #1: back injury Group #2: low protein intake Group #2: high carbohydrate intake Group #3: stress Group #4: low food intake Patient 39 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 150 152 100 126 148 158 140 150 124 126 96 152 Right: LU PC HT SI TH LI SP LR KI BL GB ST 166 126 122 128 190 154 146 142 108 90 132 192 Diagnosis: Group #1: nerve damage — upper and lower back Group #2: coronary artery disease Group #3: female Group #3: ovulation Patient 40 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 172 156 132 32 64 42 182 84 188 180 112 132 Right: LU PC HT SI TH LI SP LR KI BL GB ST 108 114 86 48 50 36 128 98 172 138 118 142 Diagnosis: Group #1: high carbohydrate intake Group #1: low protein intake Group #2: lower and upper back irritation Group #2: bladder dysfunction Group #3: stress Patient 41 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 176 98 98 86 182 148 82 198 92 66 172 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 164 98 76 94 188 126 80 188 86 96 188 192 Diagnosis: Group #1: hyperthyroid Group #2: high fat intake Group #2: spicy or high fat injestion Group #3: post nasal drip Group #4: nerve damage — upper and lower back Group #4: MS Group #5: low blood pressure Group #5: hypotension Patient 42 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 98 86 84 90 58 66 116 152 172 64 112 152 Right: LU PC HT SI TH LI SP LR KI BL GB ST 154 112 86 112 102 76 66 136 154 86 156 132 Diagnosis: Group #1: neck injury Group #1: nerve damage — lower back Group #2: hypotension Group #2: fatigue Group #2: headaches Patient 43 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 58 42 36 18 14 22 62 16 32 80 10 42 Right: LU PC HT SI TH LI SP LR KI BL GB ST 86 54 50 22 62 54 64 14 80 92 16 18 Diagnosis: Group #1: lower and upper back irritation Group #1: neck irritation Group #2: low protein intake Group #3: hyperthyroid Group #4: post nasal drip Patient 44 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 20 46 28 8 18 16 68 108 16 24 50 64 Right: LU PC HT SI TH LI SP LR KI BL GB ST 20 10 8 30 12 32 16 58 22 20 66 34 Diagnosis: Group #1: fatigue Group #1: sleep disturbance Group #1: possible coronary artery disease Group #1: microcirculatory-orthostatic hypotension Group #1: microcirculatory problem Group #1: lack of sleep Group #2: depression Group #2: inflamation Group #3: reflux Group #4: bladder dysfunction Patient 45 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 78 54 30 34 24 44 64 42 62 88 40 64 Right: LU PC HT SI TH LI SP LR KI BL GB ST 94 44 30 76 52 94 98 70 48 42 78 72 Diagnosis: Group #1: microcirculatory problem Group #1: possible coronary artery disease Group #2: lower back injury Group #2: lower back pain Group #2: lack of sleep Group #3: shoulder pain Group #4: post nasal drip Patient 46 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 106 94 72 46 82 106 86 80 100 92 102 146 Right: LU PC HT SI TH LI SP LR KI BL GB ST 112 60 86 90 88 126 60 164 54 60 200 176 Diagnosis: Group #1: possible coronary artery disease Group #2: nerve damage — upper and lower back Group #2: bladder dysfunction Group #3: shoulder pain Group #4: shoulder pain Group #5: reflux Group #6: spicy or high fact injestion Patient 47 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 158 168 162 114 170 154 166 176 156 166 120 162 Right: LU PC HT SI TH LI SP LR KI BL GB ST 158 118 122 174 154 166 156 194 150 142 166 200 Diagnosis: Group #1: possible coronary artery disease Group #2: inflamation Patient 48 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 92 64 54 82 68 112 64 114 200 80 140 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 122 68 112 64 112 174 100 146 200 38 134 200 Diagnosis: Group #1: microcirculatory problem Group #1: microcirculatory-orthostatic hypotension Group #1: pissible coronary artery disease Group #2: lower back injury Group #2: nerve damage — lower back Group #2: neck injury Group #2: neck irritation Group #3: helicobacter infection of stomach Group #4: shoulder pain Group #5: reflux Group #6: depression Patient 49 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 114 108 124 22 22 46 132 68 84 128 44 78 Right: LU PC HT SI TH LI SP LR KI BL GB ST 92 44 60 26 50 100 124 54 80 128 34 50 Diagnosis: Group #1: stress Group #2: back injury Group #3: post nasal drip Patient 50 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 134 68 38 128 128 144 46 42 42 86 26 76 Right: LU PC HT SI TH LI SP LR KI BL GB ST 54 64 60 48 136 108 42 30 62 56 48 38 Diagnosis: Group #1: low blood count Group #2: nerve damage — upper and lower back Group #2: nerve damage — neck Group #3: hyperthyroid Group #4: large bowel issues Group #5: duodenal irritation Group #6: low protein intake Group #7: sinus congestion Group #7: post nasal drip Patient 51 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 144 128 130 52 24 58 62 50 122 40 72 144 Right: LU PC HT SI TH LI SP LR KI BL GB ST 128 112 140 116 40 66 132 184 76 106 160 180 Diagnosis: Group #1: depression Group #2: hypertension Group #3: nerve damage — neck Group #3: nerve damage — lower back Group #3: bladder dysfunction Group #4: food related sinus allergy Group #4: food related stomach irritation (spicy, cheese) Group #4: reflux Patient 52 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 60 108 112 38 16 14 80 72 70 96 48 76 Right: LU PC HT SI TH LI SP LR KI BL GB ST 100 150 138 88 56 82 80 84 86 100 82 94 Diagnosis: Group #1: hyperthyroid Group #2: stress Group #2: hypertension Group #3: back injury Group #4: large bowel issues Patient 53 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 88 104 70 50 80 82 86 96 68 80 36 88 Right: LU PC HT SI TH LI SP LR KI BL GB ST 86 40 52 52 48 64 90 88 182 96 124 138 Diagnosis: Group #1: possible coronary artery disease Group #1: chronic fatigue Group #1: blood pressure medications overdose Group #1: fatigue Group #1: migraines Group #2: neck irritation Group #2: neck injury Group #2: shoulder pain Group #3: reflux Patient 54 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 92 62 64 36 24 38 110 48 80 90 32 70 Right: LU PC HT SI TH LI SP LR KI BL GB ST 112 102 96 32 34 98 92 80 56 94 76 78 Diagnosis: Group #1: post nasal drip Group #2: high stress Group #3: lower back pain Group #4: high carbohydrate intake Group #5: large bowel issues Patient 55 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 200 198 174 200 200 200 200 200 200 142 164 148 Right: LU PC HT SI TH LI SP LR KI BL GB ST 200 178 144 190 198 188 154 182 170 154 148 184 Diagnosis: Group #1: nerve damage — lower back Group #1: bladder dysfunction Group #2: neck irritation Group #3: microcirculatory problem Group #4: sinus irritation Patient 56 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 56 44 52 56 46 50 34 70 48 14 80 96 Right: LU PC HT SI TH LI SP LR KI BL GB ST 134 62 46 82 48 84 42 44 58 16 54 44 Diagnosis: Group #1: sinus irritation Group #2: duodenal irritation Group #3: stomach irritation Group #4: shoulder pain Group #4: nerve damage — lower back Group #4: bladder dysfunction Rule Outs: Group #1: lung cancer Patient 57 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 144 124 136 124 148 154 172 132 188 176 138 148 Right: LU PC HT SI TH LI SP LR KI BL GB ST 144 116 90 88 80 124 186 162 188 196 176 172 Diagnosis: Group #1: microcirculatory problem Group #2: chronic fatigue Group #3: neck injury Group #3: headaches Group #4: lower back injury Group #4: lower back pain Group #5: shoulder pain Group #6: stomach irritation Group #7: reflux Patient 58 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 70 18 22 28 54 52 112 74 98 126 38 82 Right: LU PC HT SI TH LI SP LR KI BL GB ST 62 56 48 94 62 36 88 60 68 112 38 50 Diagnosis: Group #1: microcirculatory problem Group #1: coronary artery disease Group #1: headaches Group #2: duodenal irritation Group #3: neck irritation Group #4: back injury Group #4: bladder dysfunction Group #5: microcirculatory-orthostatic hypotension related to neck, back injury Patient 59 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 110 92 120 16 70 64 50 92 10 30 30 96 Right: LU PC HT SI TH LI SP LR KI BL GB ST 54 28 32 106 160 164 46 102 12 48 22 48 Diagnosis: Group #1: depression Group #2: microcirculatory-orthostatic hypotension related to neck, back injury Group #3: nerve damage — upper and lower back Group #3: bladder dysfunction Group #4: gallbladder dysfunction Group #5: coronary artery disease Rule Outs: Group #3: MS Patient 60 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 42 84 62 58 64 60 138 116 76 148 84 104 Right: LU PC HT SI TH LI SP LR KI BL GB ST 92 80 36 70 96 122 146 150 74 130 96 184 Diagnosis: Group #1: coronary artery disease Group #1: microcirculatory problem Group #1: migraines Group #2: microcirulatory-orthostatic hypotension related to neck, back injury Group #3: stomach irritation Group #4: large bowel issues Group #4: large intestinal irritation Group #4: sinus congestion Patient 61 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 106 74 62 100 102 112 108 152 132 134 128 138 Right: LU PC HT SI TH LI SP LR KI BL GB ST 146 74 58 112 106 100 120 170 108 94 154 172 Diagnosis: Group #1: coronary artery disease Group #1: hypotension Group #1: blood pressure medications overdose Group #1: microcirculatory-orthostatic hypotension related to neck, back injury Group #2: food related stomach irritation (spicy, cheese) Group #3: post nasal drip Group #4: lower and upper back irritation Group #4: neck irritation Group #5: depression Patient 62 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 198 100 104 110 118 118 150 130 120 170 90 184 Right: LU PC HT SI TH LI SP LR KI BL GB ST 136 78 78 148 140 130 170 102 102 164 110 160 Diagnosis: Group #1: coronary artery disease Group #1: blood pressure medications overdose Group #1: microcirculatory problem Group #2: microcirculatory-orthostatic hypotension related to neck, back injury Group #3: post nasal drip Rule Outs: Group #1: headaches Group #1: migraines Group #3: lung cancer Group #3: lung disease Group #4: low back irritation Group #4: bladder dysfunction Group #5: nerve damage — neck Group #6: stomach irritation Patient 63 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 108 142 124 42 42 42 72 104 110 22 50 104 Right: LU PC HT SI TH LI SP LR KI BL GB ST 38 30 32 36 52 18 74 70 104 90 88 44 Diagnosis: Group #1: depression Group #2: coronary artery disease Group #2: chronic fatigue Group #3: neck irritation Group #4: nerve damage — lower back Group #4: bladder dysfunction Group #4: male Group #4: possible prostate enlargement Rule Outs: Group #1: depression Group #2: microcirculatory problem Group #2: coronary artery disease Patient 64 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 132 110 106 120 106 90 154 110 112 128 68 100 Right: LU PC HT SI TH LI SP LR KI BL GB ST 116 100 106 166 124 160 140 116 78 130 190 144 Diagnosis: Group #1: shoulder pain Group #2: nerve damage — neck Group #2: neck irritation Group #3: microcirculatory problem Group #3: headaches Patient 65 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 122 114 104 36 104 46 142 84 118 146 78 118 Right: LU PC HT SI TH LI SP LR KI BL GB ST 134 124 110 50 68 70 118 88 134 170 108 118 Diagnosis: Group #1: post nasal drip Group #2: neck irritation Group #2: back injury Group #3: low food intake Patient 66 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 80 44 54 36 44 32 86 62 102 132 46 34 Right: LU PC HT SI TH LI SP LR KI BL GB ST 128 84 58 120 24 28 68 68 138 148 72 60 Diagnosis: Group #1: lower and upper back irritation Group #1: bladder dysfunction Group #2: post nasal drip Group #3: duodenal irritation Group #4: high carbohydrate intake Group #5: low food intake Rule Outs: Group #1: prostate cancer Patient 67 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 76 40 20 52 98 88 96 104 82 64 124 90 Right: LU PC HT SI TH LI SP LR KI BL GB ST 72 86 74 98 70 90 60 158 96 126 152 136 Diagnosis: Group #1: lower back pain Group #1: lower back injury Group #1: bladder dysfunction Group #2: shoulder pain Group #2: upper back irritation Group #3: microcirculatory problem Group #3: coronary artery disease Group #3: headaches Group #4: depression Rule Outs: Group #3: coronary artery disease Patient 68 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 72 88 66 18 32 16 62 26 66 68 60 40 Right: LU PC HT SI TH LI SP LR KI BL GB ST 88 30 24 54 36 52 70 24 104 80 64 68 Diagnosis: Group #1: coronary artery disease Group #2: neck irritation Group #2: headaches Group #3: low back irritation Patient 69 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 80 52 82 64 60 36 152 38 160 146 138 52 Right: LU PC HT SI TH LI SP LR KI BL GB ST 42 30 80 66 76 62 134 44 142 152 138 68 Diagnosis: Group #1: microcirculatory problem Group #2: sinus congestion Group #3: bladder infection Group #4: back injury Group #5: low protein intake Group #6: low bowel movements Patient 70 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 98 74 46 14 26 30 104 200 68 106 96 192 Right: LU PC HT SI TH LI SP LR KI BL GB ST 136 70 46 24 16 28 86 82 80 142 100 98 Diagnosis: Group #1: sinus congestion Group #2: sinus allergy Group #3: coronary artery disease Group #4: high carbohydrate intake Group #5: allergy to food Group #6: neck irritation Group #7: migraines Patient 71 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 140 120 86 112 120 116 82 72 84 74 146 100 Right: LU PC HT SI TH LI SP LR KI BL GB ST 162 132 118 154 130 102 76 78 130 80 118 146 Diagnosis: Group #1: sinus congestion Group #2: hyperthyroid Group #3: nerve damage — upper and lower back Group #3: bladder dysfunction Group #4: neck irritation Group #5: shoulder pain Group #6: low protein intake Group #7: coronary artery disease Group #8: reflux Rule Outs: Group #1: lung cancer Group #1: lung disease Patient 72 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 130 118 112 88 130 128 156 48 54 130 88 68 Right: LU PC HT SI TH LI SP LR KI BL GB ST 146 110 108 148 148 158 130 44 78 138 150 66 Diagnosis: Group #1: weight loss Group #1: low protein intake Group #1: low food intake Group #2: low protein intake Group #3: never damage — neck Group #3: neck injury Group #4: low back irritation Group #5: shoulder pain Group #6: post nasal drip Patient 73 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 96 72 58 114 142 146 86 66 60 68 52 92 Right: LU PC HT SI TH LI SP LR KI BL GB ST 94 70 60 80 92 108 78 46 76 56 44 94 Diagnosis: Group #1: large bowel issues Group #2: hyperthyroid Group #3: nerve damage — upper and lower back Group #4: low food intake Group #5: coronary artery disease Group #6: headaches Patient 74 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 168 152 154 170 166 186 180 186 168 188 80 148 Right: LU PC HT SI TH LI SP LR KI BL GB ST 180 124 126 160 142 158 140 98 152 182 92 90 Diagnosis: Group #1: lower back pain Group #1: lower back injury Group #1: bladder dysfunction Group #2: post nasal drip Group #2: allergy Group #3: large bowel issues Patient 75 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 64 70 88 136 172 194 136 148 170 114 88 168 Right: LU PC HT SI TH LI SP LR KI BL GB ST 150 92 86 178 168 190 136 96 172 82 164 200 Diagnosis: Group #1: low blood pressure Group #1: dysautonomia Group #1: neck irritation Group #1: headaches Group #1: migraines Group #2: nerve damage — lower back Group #2: bladder dysfunction Group #3: shoulder pain Group #3: reflux Group #3: stomach irritation Group #4: sinus congestion Patient 76 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 152 134 70 98 132 156 76 80 68 82 42 46 Right: LU PC HT SI TH LI SP LR KI BL GB ST 154 48 30 116 110 110 94 40 50 72 22 68 Diagnosis: Group #1: orthostatic hypotension Group #1: migraines Group #2: nerve damage — neck Group #3: post nasal drip Group #4: microcirculatory problem Group #4: microcirculatory-orthostatic hypotension related to neck, back injury Group #5: weight loss Group #5: low food intake Patient 77 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 134 158 106 108 110 170 80 76 88 42 74 54 Right: LU PC HT SI TH LI SP LR KI BL GB ST 138 72 70 102 120 160 86 44 70 80 72 62 Diagnosis: Group #1: coronary artery disease Group #1: migraines Group #2: nerve damge — upper and lower back Group #2: bladder dysfunction Group #3: large bowel issues Group #4: hyperthyroid Group #5: post nasal drip Group #6: microcirculatory problem Patient 78 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 114 108 106 106 94 94 62 126 70 74 62 98 Right: LU PC HT SI TH LI SP LR KI BL GB ST 118 72 76 120 88 50 80 82 70 72 96 140 Diagnosis: Group #1: coronary artery disease Group #2: nerve damage — upper and lower back Group #2: bladder dysfunction Group #3: depression Group #4: post nasal drip Group #5: food related stomach irritation (spicy, cheese) Patient 79 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 120 98 104 26 40 60 116 90 92 160 124 98 Right: LU PC HT SI TH LI SP LR KI BL GB ST 130 128 88 80 68 54 120 54 104 140 106 88 Diagnosis: Group #1: lower back pain Group #1: lower back injury Group #2: post nasal drip Group #3: shoulder pain Group #3: upper back irritation Rule Outs: Group #2: lung disease Patient 80 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 156 146 130 94 104 118 174 162 196 138 94 140 Right: LU PC HT SI TH LI SP LR KI BL GB ST 182 168 174 100 70 126 182 172 160 138 128 172 Diagnosis: Group #1: neck injury Group #1: neck irritation Group #2: depression Patient 81 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 152 86 94 110 116 118 76 174 110 92 88 110 Right: LU PC HT SI TH LI SP LR KI BL GB ST 152 82 84 42 42 78 76 182 68 70 122 124 Diagnosis: Group #1: post nasal drip Group #1: allergy Group #2: coronary artery disease Group #3: nerve damage — neck Group #3: nerve damage — lower back Group #4: microcirculatory-orthostatic hypotension Group #4: migraines Group #4: bladder dysfunction Patient 82 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 162 146 154 134 136 118 100 200 114 98 200 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 200 200 142 182 108 112 120 200 124 80 188 200 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: nerve damage — neck Group #1: bladder dysfunction Group #2: microcirculatory problem Group #3: food related stomach irritation (spicy, cheese) Group #3: spicy or high fat injestion Group #4: post nasal drip Patient 83 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 120 118 142 134 120 92 146 184 188 182 174 176 Right: LU PC HT SI TH LI SP LR KI BL GB ST 180 108 130 132 130 114 158 184 190 170 186 192 Diagnosis: Group #1: microcirculatory-orthostatic hypotension Group #1: fatigue Group #1: coronary artery disease Group #1: low blood pressure Group #2: post nasal drip Group #2: sinus congestion Group #3: neck irritation Group #3: lower back pain Patient 84 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 100 144 162 96 116 132 84 90 60 108 84 140 Right: LU PC HT SI TH LI SP LR KI BL GB ST 108 98 154 140 130 132 134 126 66 74 144 148 Diagnosis: Group #1: microcirculatory problem Group #2: stress Group #3: nerve damage — upper and lower back Group #3: nerve damage — neck Group #3: bladder dysfunction Group #4: reflux Group #6: sinus congestion Patient 85 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 72 82 70 102 158 126 78 102 90 48 90 122 Right: LU PC HT SI TH LI SP LR KI BL GB ST 152 162 130 164 164 102 32 112 82 44 98 102 Diagnosis: Group #1: bladder dysfunction Group #1: nerve damage — lower back Group #1: nerve damage — neck Group #2: hyperthyroid Group #3: stress Group #4: microcirculatory problem Group #5: sinus congestion Patient 86 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 132 98 124 136 132 146 122 114 92 80 112 172 Right: LU PC HT SI TH LI SP LR KI BL GB ST 114 116 104 152 162 154 98 114 86 74 138 128 Diagnosis: Group #1: nerve damage — lower back Group #1: nerve damage — neck Group #2: fatigue Rule Outs: Group #1: MS Group #1: bladder dysfunction Group #2: coronary artery disease Group #2: reflux Group #3: hyperthyroid Patient 87 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 136 106 136 74 58 50 170 94 122 168 158 116 Right: LU PC HT SI TH LI SP LR KI BL GB ST 96 92 124 70 108 48 160 122 170 128 138 108 Diagnosis: Group #1: fatigue Group #1: neck irritation Group #1: back injury Group #1: shoulder pain Patient 88 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 102 124 134 88 68 66 46 56 50 44 22 44 Right: LU PC HT SI TH LI SP LR KI BL GB ST 62 64 62 102 90 94 14 60 50 20 50 64 Diagnosis: None Rule Outs: Group #1: MS Group #1: nerve damage — lower back Group #1: nerve damage — neck Group #1: bladder dysfunction Group #2: stress Group #3: duodenal irritation Patient 89 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 78 38 50 78 66 60 64 14 24 24 30 18 Right: LU PC HT SI TH LI SP LR KI BL GB ST 86 62 46 68 26 54 18 30 18 16 34 28 Diagnosis: Group #1: post nasal drip Group #2: duodenal irritation Group #3: nerve damage — neck Group #3: nerve damage — lower back Group #3: bladder dysfunction Rule Outs: Group #1: post nasal drip Group #1: lung cancer Group #1: lung disease Patient 90 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 100 68 72 54 36 58 106 78 102 128 76 80 Right: LU PC HT SI TH LI SP LR KI BL GB ST 90 76 82 50 180 76 130 86 104 106 90 98 Diagnosis: Group #1: back injury Group #1: bladder dysfunction Rule Outs: Group #2: thyroid problem Group #2: pituitary problem Patient 91 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 116 78 84 46 64 52 146 64 110 100 72 80 Right: LU PC HT SI TH LI SP LR KI BL GB ST 94 98 74 82 84 42 138 72 120 122 96 62 Diagnosis: Group #1: back injury Group #1: neck irritation Group #2: large bowel issues Patient 92 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 158 158 150 170 134 156 118 116 104 98 158 152 Right: LU PC HT SI TH LI SP LR KI BL GB ST 148 136 140 194 170 172 94 146 94 92 184 170 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: bladder dysfunction Group #2: shoulder pain Group #2: reflux Group #3: duodenal irritation Group #3: stress Patient 93 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 42 52 52 44 52 24 48 76 84 52 102 144 Right: LU PC HT SI TH LI SP LR KI BL GB ST 78 36 36 70 42 68 54 138 80 70 180 158 Diagnosis: Group #1: nerve damage — lower back Group #1: bladder dysfunction Group #2: shoulder pain Group #3: reflux Group #4: sinus allergy Group #4: sinus congestion Group #5: stomach irritation Patient 94 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 62 16 16 52 94 104 8 12 22 16 16 18 Right: LU PC HT SI TH LI SP LR KI BL GB ST 112 32 24 60 86 116 22 18 14 20 22 18 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: nerve damage — neck Group #2: microcirculatory-orthostatic hypotension related to neck, back injury Group #2: microcirculatory problem Group #3: post nasal drip Group #4: large bowel issues Group #4: large intestinal irritation Group #5: thyroid problem Rule Outs: Group #6: gastroparesis Group #6: MS Group #6: bladder dysfunction Patient 95 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 176 156 156 10 24 26 200 186 186 188 190 188 Right: LU PC HT SI TH LI SP LR KI BL GB ST 176 182 144 56 52 42 200 162 200 200 186 142 Diagnosis: Group #1: back injury Group #1: neck injury Group #1: neck irritation Group #1: lower and upper back irritation Group #2: shoulder pain Group #3: reflux Group #4: food related stomach irritation (spicy, cheese) Group #5: high carbohydrate intake Group #6: food related sinus allergy Group #6: post nasal drip Patient 96 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 138 130 120 154 148 124 88 98 128 114 92 108 Right: LU PC HT SI TH LI SP LR KI BL GB ST 170 128 152 160 140 96 120 102 106 66 96 120 Diagnosis: Group #1: nerve damage — lower back Group #2: duodenal irritation Group #3: post nasal drip Group #4: stress Group #5: hyperthyroid Patient 97 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 180 144 142 64 16 40 104 36 46 132 58 52 Right: LU PC HT SI TH LI SP LR KI BL GB ST 150 102 116 70 26 48 136 44 28 128 14 66 Diagnosis: Group #1: high carbohydrate intake Group #1: low protein intake Group #2: high stress Group #3: post nasal drip Group #4: lower back injury Rule Outs: Group #4: bladder dyfunction Patient 98 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 132 158 138 98 68 98 110 62 74 60 50 90 Right: LU PC HT SI TH LI SP LR KI BL GB ST 162 154 76 52 72 70 82 30 100 94 40 92 Diagnosis: Group #1: high stress Group #2: nerve damage — neck Group #2: nerve damage — lower back Group #2: neck irritation Group #3: low protein intake Patient 99 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 178 162 174 178 186 162 196 196 146 194 164 178 Right: LU PC HT SI TH LI SP LR KI BL GB ST 148 136 140 146 180 156 188 194 148 196 170 172 Diagnosis: Group #1: neurodeficiency due to to neck injury Group #1: microcirculatory problem Rule Outs: Group #1: coronary artery disease Group #2: nerve damage Group #3: low back irritation Group #3: lower back pain Group #3: lower back injury Patient 100 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 138 88 132 76 78 50 186 106 162 178 178 124 Right: LU PC HT SI TH LI SP LR KI BL GB ST 100 88 132 120 74 44 162 74 156 164 0 0 Diagnosis: Group #1: lower back injury Group #1: neck injury Group #3: post nasal drip Group #4: shoulder pain Group #4: reflux Rule Outs: Group #1: bladder dysfunction Group #2: microcirculatory problem Group #2: headaches Group #2: migraines Group #3: post nasal drip Patient 101 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 134 104 152 174 148 82 186 80 146 174 174 122 Right: LU PC HT SI TH LI SP LR KI BL GB ST 48 44 124 134 140 66 180 106 164 172 178 90 Diagnosis: Group #1: microcirculatory problem Group #2: reflux Group #2: sinus congestion Group #3: neck irritation Group #3: low back irritation Rule Outs: Group #2: esophageal obstruction Group #2: sinus infection Group #3: nerve related slow peristaltic activity Group #3: bladder dysfunction Group #4: stress Patient 102 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 122 124 122 92 76 148 170 102 110 148 132 182 Right: LU PC HT SI TH LI SP LR KI BL GB ST 116 138 146 106 108 104 154 144 106 146 108 178 Diagnosis: Group #1: stress Group #1: nerve damage — neck Group #1: back injury Group #2: stomach irritation Rule Outs: Group #1: depression Group #3: large bowel issues Patient 103 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 22 34 34 60 66 60 84 136 94 96 110 90 Right: LU PC HT SI TH LI SP LR KI BL GB ST 68 56 62 76 72 70 82 136 98 70 128 126 Diagnosis: Group #1: low blood pressure Group #2: food related stomach irritation (spicy, cheese) Group #3: sinus congestion Group #4: neck irritation Group #4: low back irritation Group #5: shoulder pain Rule Outs: Group #1: blood pressure medications overdose Group #2: allergy to food Patient 104 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 94 50 74 54 96 166 66 110 62 52 154 126 Right: LU PC HT SI TH LI SP LR KI BL GB ST 172 66 56 136 172 112 56 26 44 44 90 112 Diagnosis: Group #1: low blood pressure Group #1: orthostatic hypotension Group #2: nerve damage — lower back Group #2: nerve damage — neck Group #2: bladder dysfunction Group #4: shoulder pain Group #7: duodenal irritation Group #8: reflux Rule Outs: Group #2: MS Group #4: depression Group #5: thyroid problem Group #6: thyroid problem Group #7: large bowel issues Patient 105 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 118 144 132 72 68 114 126 162 156 84 128 166 Right: LU PC HT SI TH LI SP LR KI BL GB ST 146 144 118 100 120 114 74 170 176 144 148 176 Diagnosis: Group #1: neck irritation Group #2: nerve damage — lower back Group #3: thyroid problem Group #4: stomach irritation Group #5: shoulder pain Group #6: post nasal drip Group #7: food related stomach irritation (spicy, cheese) Rule Outs: Group #2: bladder dysfunction Patient 106 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 144 130 136 104 128 108 158 140 50 156 136 138 Right: LU PC HT SI TH LI SP LR KI BL GB ST 174 142 158 160 142 144 162 180 40 124 176 134 Diagnosis: Group #1: nerve damage — neck Group #2: back injury Group #3: nerve damage — neck Group #3: neck injury Group #4: stress Group #5: post nasal drip Group # Rule Outs: Group #5: lung disease Patient 107 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 158 138 130 52 76 88 100 80 90 140 46 100 Right: LU PC HT SI TH LI SP LR KI BL GB ST 158 116 112 164 166 114 110 74 92 128 76 66 Diagnosis: Group #1: nerve damage — neck Group #2: back injury Group #3: post nasal drip Group #4: stress Group #5: nerve damage — neck Group #6: low food intake Rule Outs: Group #2: male Group #2: prostate cancer Group #2: bladder dysfunction Group #7: thyroid problem Patient 108 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 44 56 60 132 134 112 70 22 74 64 16 14 Right: LU PC HT SI TH LI SP LR KI BL GB ST 86 74 46 138 84 86 80 20 58 52 68 42 Diagnosis: Group #1: weight loss Group #1: low food intake Group #1: low protein intake Group #2: sinus congestion Group #2: post nasal drip Patient 109 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 62 72 28 68 68 54 108 96 88 78 56 126 Right: LU PC HT SI TH LI SP LR KI BL GB ST 50 50 50 78 46 54 94 92 80 62 104 108 Diagnosis: Group #1: low blood pressure Group #2: sinus infection Group #3: reflux Group #3: irritation of stomach lining Group #4: large bowel issues Rule Outs: Group #1: coronary artery disease Group #2: sinus infection Group #2: infection bacterial Group #3: infection bacterial Group #5: hypothyroid disease Patient 110 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 130 164 158 44 60 58 146 132 146 158 134 150 Right: LU PC HT SI TH LI SP LR KI BL GB ST 122 116 110 24 46 82 170 146 126 162 144 130 Diagnosis: Group #1: depression Group #2: high carbohydrate intake Group #3: low back irritation Group #3: neck irritation Group #6: infection bacterial Rule Outs: Group #3: bladder dysfunction Group #4: shoulder pain Group #5: reflux Patient 111 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 144 98 70 44 104 54 54 108 50 50 106 178 Right: LU PC HT SI TH LI SP LR KI BL GB ST 82 28 40 56 30 100 48 120 50 62 98 166 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: nerve damage — neck Group #3: dysautonomia Group #4: stomach irritation Group #5: post nasal drip Group #6: reflux Group #7: food related stomach irritation (spicy, cheese) Group #8: large bowel issues Rule Outs: Group #2: coronary artery disease Group #3: orthostatic hypotension Patient 112 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 118 122 150 146 108 142 46 156 182 96 92 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 134 154 154 182 178 180 102 86 156 60 80 184 Diagnosis: Group #1: nerve damage — lower back Group #2: neck injury Group #3: stomach irritation Group #4: depression Group #5: sinus allergy Group #6: stress Group #7: large bowel issues Rule Outs: Group #6: thyroid problem Patient 113 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 136 108 98 120 76 98 102 90 104 108 70 118 Right: LU PC HT SI TH LI SP LR KI BL GB ST 144 112 98 148 132 104 76 116 100 124 84 76 Diagnosis: Group #1: low blood count Group #1: high blood sugar Group #2: post nasal drip Group #3: lower back injury Group #3: nerve damage — lower back Group #4: post nasal drip Rule Outs: Group #2: lung cancer Group #2: lung disease Group #4: lung cancer Group #4: lung disease Patient 114 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 160 160 132 86 120 136 176 56 152 200 106 98 Right: LU PC HT SI TH LI SP LR KI BL GB ST 132 94 74 150 162 156 132 80 86 146 166 100 Diagnosis: Group #1: nerve damage — neck Group #1: neck irritation Group #2: migraines Group #2: microcirculatory problem Group #4: back injury Group #6: shoulder pain Group #7: post nasal drip Group #8: low food intake Group #8: low protein intake Rule Outs: Group #3: coronary artery disease Group #4: bladder dysfunction Group #5: male Group #5: prostate cancer Patient 115 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 136 92 114 82 134 126 116 120 138 146 132 124 Right: LU PC HT SI TH LI SP LR KI BL GB ST 138 82 94 72 76 118 136 140 140 166 104 132 Diagnosis: Group #1: migraines Group #1: fatigue Group #2: neck irritation Group #2: back injury Group #3: orthostatic hypotension Rule Outs: Group #1: coronary artery disease Group #3: bladder dysfunction Group #4: sinus allergy Group #4: post nasal drip Patient 116 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 166 136 120 184 182 184 46 84 198 46 58 148 Right: LU PC HT SI TH LI SP LR KI BL GB ST 160 136 130 182 184 180 70 94 92 66 118 116 Diagnosis: Group #1: weight loss Group #1: low protein intake Group #2: post nasal drip Group #2: nerve damage — lower back Group #2: bladder dysfunction Group #3: neck irritation Group #3: nerve damage — neck Rule Outs: Group #1: weight loss Group #2: lung cancer Group #2: lung disease Patient 117 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 124 108 88 162 136 182 104 76 98 108 80 76 Right: LU PC HT SI TH LI SP LR KI BL GB ST 200 158 106 178 188 174 118 122 102 130 130 126 Diagnosis: Group #1: microcirculatory problem Group #1: coronary artery disease Group #2: post nasal drip Group #3: nerve damage — neck Group #4: nerve damage — lower back Rule Outs: Group #1: coronary artery disease Group #4: bladder dysfunction Group #5: thyroid problem Group #6: large bowel issues Group #7: duodenal irritation Group #8: weight loss Patient 118 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 118 96 94 30 56 52 86 86 86 98 56 78 Right: LU PC HT SI TH LI SP LR KI BL GB ST 134 80 80 98 86 64 108 76 82 88 90 74 Diagnosis: Group #1: post nasal drip Rule Outs: Group #1: lung cancer Group #1: lung disease Patient 119 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 186 174 162 86 100 90 124 88 200 112 132 124 Right: LU PC HT SI TH LI SP LR KI BL GB ST 170 100 150 138 136 88 144 100 188 126 160 116 Diagnosis: Group #1: microcirculatory problem Group #2: stress Group #3: post nasal drip Group #4: neck injury Group #4: neck irritation Group #5: low protein intake Group #6: nerve damage — lower back Rule Outs: Group #6: bladder dysfunction Group #7: large bowel issues Group #8: thyroid problem Patient 120 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 136 162 82 38 40 72 116 152 94 124 180 200 Right: LU PC HT SI TH LI SP LR KI BL GB ST 146 138 106 84 118 64 144 200 88 106 172 200 Diagnosis: Group #1: food related stomach irritation (spicy, cheese) Group #2: reflux Group #3: nerve damage — neck Group #6: post nasal drip Rule Outs: Group #4: thyroid problem Group #5: slow duodenal peristalsis Group # Patient 121 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 106 70 54 106 136 44 158 94 64 136 124 80 Right: LU PC HT SI TH LI SP LR KI BL GB ST 118 106 52 194 136 68 162 72 40 132 178 90 Diagnosis: Group #3: duodenal irritation Group #4: microcirculatory-orthostatic hypotension related to neck, back injury Group #5: nerve damage — neck Group #6: back injury Rule Outs: Group #1: pituitary problem Group #1: thyroid problem Group #6: male Group #6: prostate cancer Group #6: bladder dysfunction Group #7: shoulder pain Group #8: large bowel issues Group #9: infection bacterial Patient 122 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 90 94 108 46 54 108 132 86 132 128 102 132 Right: LU PC HT SI TH LI SP LR KI BL GB ST 100 100 86 114 82 92 120 92 104 104 104 96 Diagnosis: Group #1: neck irritation Group #1: back injury Group #2: stomach irritation Group #3: slow duodenal peristalsis Rule Outs: Group #1: thyroid problem Patient 123 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 110 84 72 168 110 180 134 36 104 132 36 40 Right: LU PC HT SI TH LI SP LR KI BL GB ST 110 100 92 120 96 122 98 24 78 112 80 76 Diagnosis: Group #1: low food intake Group #1: low protein intake Group #2: duodenal irritation Group #3: large intestinal irritation Group #4: back injury Group #5: large bowel issues Group #6: duodenal irritation Rule Outs: Group #4: coronary artery disease Patient 124 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 158 88 114 166 196 174 120 94 108 118 100 102 Right: LU PC HT SI TH LI SP LR KI BL GB ST 184 146 150 150 176 198 146 100 130 132 130 134 Diagnosis: Group #1: nerve damage — upper and lower back Group #1: nerve damage — neck Group #2: microcirculatory problem Group #3: low food intake Group #3: low protein intake Group #3: weight loss Group #3: post nasal drip Rule Outs: Group #1: bladder dysfunction Group #2: coronary artery disease Patient 125 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 118 106 108 26 32 40 36 94 54 26 12 142 Right: LU PC HT SI TH LI SP LR KI BL GB ST 108 84 90 24 82 182 56 64 76 38 56 68 Diagnosis: Group #1: nerve damage — lower back Group #1: nerve damage — neck Group #2: stomach irritation Group #3: large bowel issues Group #3: large intestinal irritation Group #4: depression Group #5: sinus allergy Group #5: post nasal drip Patient 126 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 98 70 84 68 110 130 108 108 36 36 76 60 Right: LU PC HT SI TH LI SP LR KI BL GB ST 88 76 54 114 90 118 102 106 64 60 94 88 Diagnosis: Group #1: nerve damage — lower back Group #1: nerve damage — neck Group #3: thyroid problem Rule Outs: Group #2: large intestinal irritation Group #2: autoimmune disease Group #4: bladder dysfunction Patient 127 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 196 190 178 114 186 148 82 120 84 100 88 70 Right: LU PC HT SI TH LI SP LR KI BL GB ST 200 150 180 180 176 164 112 70 80 136 106 88 Diagnosis: Group #2: post nasal drip Group #3: weight loss Group #3: low food intake Group #3: low protein intake Group #4: hyperthyroid Group #5: nerve damage — upper and lower back Group #5: nerve damage — neck Rule Outs: Group #1: stress Group #2: lung cancer Group #2: lung disease Group #4: hyperthyroid Group #5: MS Group #5:bladder dysfunction Patient 128 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 128 72 112 62 50 78 80 58 72 110 8 58 Right: LU PC HT SI TH LI SP LR KI BL GB ST 104 76 64 82 70 54 98 92 84 82 36 78 Diagnosis: Group #1: post nasal drip Group #1: allergy Group #2: depression Group #3: low back irritation Group #4: gallbladder dysfunction Group #4: gallbladder problem Group #6: constipation Rule Outs: Group #5: thyroid problem Patient 129 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 92 24 20 24 78 142 48 38 50 38 178 198 Right: LU PC HT SI TH LI SP LR KI BL GB ST 148 52 78 78 50 146 92 168 152 50 200 200 Diagnosis: Group #1: microcirculatory-orthostatic hypotension related to neck, back injury Group #1: microciruclatory problem Group #2: nerve damage Group #2: nerve damage — lower back Group #2: nerve damage — neck Group #2: neck irritation Group #3: bladder dysfunction Group #5: shoulder pain Group #6: reflux Group #6: stomach irritation Group #7: food related stomach irritation (spicy, cheese) Group #8: sinus allergy Group #8: post nasal drip Rule Outs: Group #2: MS Patient 130 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 162 102 126 90 82 114 64 88 122 92 126 114 Right: LU PC HT SI TH LI SP LR KI BL GB ST 192 88 96 96 96 128 200 64 116 96 154 148 Diagnosis: Group #3: lower back injury Group #4: lower back pain Rule Outs: Group #1: coronary artery disease Group #2: lung cancer Group #2: lung disease Symptoms: Group #2: post nasal drip Group #4: lower back pain Group #4: lower back injury Patient 131 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 78 70 36 146 106 150 74 110 70 170 58 168 Right: LU PC HT SI TH LI SP LR KI BL GB ST 96 52 54 122 40 116 66 68 78 96 92 188 Diagnosis: Group #3: nerve damage — neck Group #3: neck injury Group #4: lower back injury Group #4: lower back pain Group #5: microcirculatory-orthostatic hypotension Group #6: stomach irritation Group #7: duodenal irritation Rule Outs: Group #2: coronary artery disease Symptoms: Group #1: low blood pressue Patient 132 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 54 20 30 90 62 122 48 42 14 32 42 66 Right: LU PC HT SI TH LI SP LR KI BL GB ST 48 20 24 40 70 50 36 62 26 22 84 58 Diagnosis: Group #1: neurodeficiency due to neck injuyr Group #1: orthostatic hypotension Group #1: migraines Group #4: shoulder pain Group #5: nerve damage — neck Group #6: duodenal irritation Rule Outs: Group #2: coronary artery disease Group #3: constipation Patient 133 Measurements: Left: LU PC HT SI TH LI SP LR KI BL GB ST 120 132 122 30 126 120 136 142 118 148 122 110 Right: LU PC HT SI TH LI SP LR KI BL GB ST 140 122 106 74 102 82 128 96 82 128 126 120 Diagnosis: Group #1: lower back injury Group #1: lower back pain Group #2: nerve damage — neck Group #3: bladder dysfunction

Variations

This invention may be implemented in many different ways. Here are some non-limiting examples:

In some implementations, this invention is a method comprising: (a) taking, in a diagnostic session, a set of measurements of electric current that flow between a ground electrode and a probe electrode through a patient’s body, the measurements being taken in such a way that (i) different measurements in the set are taken while the probe electrode touches skin of the patient at different Prototype Measurement Locations, one location at a time, and (ii) each of the respective measurements in the set is taken while (A) the ground electrode touches skin of a hand of a forearm of the patient, and (B) the probe electrode touches skin of another limb of the patient at one of the Prototype Measurement Locations; (b) calculating, based on the set of measurements, a Prototype Electrical Current State for the diagnostic session; (c) employing a lookup table to identify a medical condition that the lookup table Associates with the Prototype Electrical Current State, which medical condition is a Prototype Medical Condition; and (d) outputting (i) a diagnosis that the patient has the medical condition, or (ii) a recommendation that the patient undergo medical testing to evaluate whether the patient has the medical condition. In some cases, the Prototype Medical Condition is a Class B Condition. In some cases, the Prototype Medical Condition is a Class M Condition. In some cases, the Prototype Medical Condition is a Class N Condition. In some cases, the Prototype Medical Condition is a Class P Condition. In some cases, the Prototype Medical Condition is a Class A Condition. In some cases, the Prototype Medical Condition is a Class C Condition. In some cases, the Prototype Medical Condition is a viral infection. In some cases, the Prototype Medical Condition is a bacterial infection. In some cases, the Prototype Medical Condition is a Class D Condition. Each of the cases described above in this paragraph is an example of the method described in the first sentence of this paragraph, and is also an example of an embodiment of this invention that may be combined with other embodiments of this invention.

In some implementations, this invention is a method comprising: (a) calculating a Prototype Electrical Current State for a diagnostic session, based on a set of measurements of electric current in a patient; (b) employing a lookup table to identify a medical condition that the lookup table Associates with the Prototype Electrical Current State, which medical condition is a Prototype Medical Condition; and (c) outputting (i) a diagnosis that the patient has the medical condition, or (ii) a recommendation that the patient undergo medical testing to evaluate whether the patient has the medical condition. In some cases, different measurements in the set were taken while: (a) the probe electrode touched skin of the patient at different Prototype Measurement Locations, one location at a time; and (b) the electric current flowed between a ground electrode and a probe electrode through the patient’s body. In some cases, each of the respective measurements in the set was taken while: (a) the electric current flowed between a ground electrode and a probe electrode through the patient’s body; (b) the ground electrode touched skin of a hand of a forearm of the patient; and (c) the probe electrode touched skin of another limb of the patient at a Prototype Measurement Location. In some cases, the Prototype Medical Condition is a Class B Condition, Class M Condition, Class N Condition or Class P Condition. Each of the cases described above in this paragraph is an example of the method described in the first sentence of this paragraph, and is also an example of an embodiment of this invention that may be combined with other embodiments of this invention.

In some implementations, this invention is a system comprising: (a) a current sensor that includes a ground electrode and a probe electrode; and (b) one or more computers; wherein (i) the current sensor is configured to take, during a diagnostic session, a set of measurements of electric current, in such a way that (A) the electric current being measured flows between the ground electrode and the probe electrode through a patient’s body, (B) different measurements in the set are taken while the probe electrode touches skin of the patient at different Prototype Measurement Locations, one location at a time, and (C) each of the respective measurements in the set is taken while (I) the ground electrode touches skin of a hand of a forearm of the patient, and (II) the probe electrode touches skin of another limb of the patient at one of the Prototype Measurement Locations, and (ii) the one or more computers are programmed (A) to calculate, based on the set of measurements, a Prototype Electrical Current State for the diagnostic session, (B) to employ a lookup table to identify a medical condition that the lookup table Associates with the Prototype Electrical Current State, which medical condition is a Prototype Medical Condition, and (C) to output (I) a diagnosis that the patient has the medical condition, or (II) a recommendation that the patient undergo medical testing to evaluate whether the patient has the medical condition. In some cases, the ground electrode and the probe electrode are parts of a single rigid structure and are in a fixed position relative to each other. In some cases: (a) the system further comprises one or more pressure sensors that are each configured to measure pressure exerted on the ground electrode or the probe electrode; and (b) the ground electrode and the probe electrode are in fixed positions relative to each other, except for any movement due to displacement that occurs within the one or more pressure sensors. In some cases: (a) the system further comprises an electronic display screen and an audio transducer; and (b) the one or more computers are programmed to cause the screen and the audio transducer to together output an audiovisual presentation that provides information about whether the ground and probe electrodes are positioned correctly on the patient. In some cases: (a) the ground electrode and the probe electrode are parts of a single rigid structure and are in a fixed position relative to each other; and (b) the rigid structure is configured to partially surround a smartphone or other mobile computing device. In some cases, the Prototype Medical Condition is a Class B Condition, Class M Condition, Class N Condition or Class P Condition. Each of the cases described above in this paragraph is an example of the system described in the first sentence of this paragraph, and is also an example of an embodiment of this invention that may be combined with other embodiments of this invention.

Each description herein (or in the Provisional) of any method, apparatus or system of this invention describes a non-limiting example of this invention. This invention is not limited to those examples, and may be implemented in other ways.

Each description herein (or in the Provisional) of any prototype of this invention describes a non-limiting example of this invention. This invention is not limited to those examples, and may be implemented in other ways.

Each description herein (or in the Provisional) of any implementation, embodiment or case of this invention (or any use scenario for this invention) describes a non-limiting example of this invention. This invention is not limited to those examples, and may be implemented in other ways.

Each Figure, diagram, schematic or drawing herein (or in the Provisional) that illustrates any feature of this invention shows a non-limiting example of this invention. This invention is not limited to those examples, and may be implemented in other ways.

The above description (including without limitation any attached drawings and figures) describes illustrative implementations of the invention. However, the invention may be implemented in other ways. The methods and apparatus which are described herein are merely illustrative applications of the principles of the invention. Other arrangements, methods, modifications, and substitutions by one of ordinary skill in the art are also within the scope of the present invention. Numerous modifications may be made by those skilled in the art without departing from the scope of the invention. Also, this invention includes without limitation each combination and permutation of one or more of the items (including any hardware, hardware components, methods, processes, steps, software, algorithms, features, and technology) that are described herein.

Non-Human Animals

The embodiments of the invention as described above may also be employed with respect to the “cross-body” electrical currents of non-human animals (i.e. dog, cat, horse, cows etc.), especially mammals. Preferably, measurement methodology for no-human mammals is substantially the same as with human patients described above, that is measurements are taken of cross-currents that flow through the animal torso between animal limbs along acupuncture meridians.

For example, measurement data is preferably taken while by probing each animal leg, with 6 locations of probing on each leg, as shown in FIGS. 14A-D with reference to examples of dogs, cats, horses and cows. Locations 1701-1706 are associated with the animal’s back legs and locations 1801-1806 are associated with the animal’s front legs.

Specifically, locations 1701-1706 are positioned on the Spleen, Liver, Kidney, Bladder, Gall Bladder, and Stomach acupuncture meridians, respectively. In acupuncture terminology: (a) location 1701 is sometimes called SP3 or Spleen 3; (b) location 1702 is sometimes called LR3 or Liver 3; (c) location 1703 is sometimes called KI4 or Kidney 4; (d) location 1704 is sometimes called BL65 or Bladder 65; (e) location 1705 is sometimes called GB40 or Gall Bladder 40; and (f) location 1706 is sometimes called ST42 or Stomach 42. Locations 1801-1806 are positioned on the Lung, Pericardium, Heart, Small Intestine, Triple Heater and Large Intestine acupuncture meridians, respectively. In acupuncture terminology: (a) location 1801 is sometimes called LU9 or Lung 9; (b) location 1802 is sometimes called PC7 or Pericardium 7; (c) location 1803 is sometimes called HT7 or Heart 7; (d) location 1804 is sometimes called SI5 or Small Intestine 5; (e) location 1805 is sometimes called TH4 or Triple Heater 4; and (f) location 1806 is sometimes called LI5 or Large Intestine 5.

These locations 1701-1706, 1801-1806 may be determined by known acupressure meridian charts of the animal species or may be determined and/or varied on a case-by-case basis by the health-care worker. Locations 1701-1706, 1801-1806 may vary by animal species and even in individual animals within a species, thus determination of the locations 1701-1706, 1801-1806 should be left to a human diagnostician. In some cases, the measurement locations may be placed on different acupuncture points than described or somewhere else on the animal that is not along any acupuncture points at all, based on diagnostician discretion. In addition, in some cases, less than 24 measurement locations are utilized during a single diagnostic session, which may also be at diagnostician discretion.

As seen in FIGS. 14A-D, three locations are preferably determined on both the inside and the outside of each animal limb. These determined locations may be positioned on the same acupuncture meridians—- and have the same acupuncture point numbers - as the respective corresponding locations on the opposite limb. For instance, FIG. 14A depicts the example of a canine. Measurement locations 1801, 1802, and 1806 are generally located laterally on (the outside of) a canine front leg, here shown on the lateral surface of a canine front left leg. Measurement locations 1804-1805 are generally located medially on (the inside of) a canine front leg, here shown on the medial surface of a canine front right leg. However, the measurement locations of the front legs preferably have substantial bilateral symmetry, thus measurement locations 1801, 1802, and 1806 may be determined for the canine front right leg and measurement locations 1803-1805 may be determined for the canine front left leg as well in mirrored positions. Thus, each limb will preferably have 6 measurement locations 1701-1706 or 1801-1806 for the diagnostician to locate and/or determine and probe, and each animal will preferably have 24 probe locations total, as was described for human patients above.

In some embodiments and/or uses of the method according to the present invention, diagnostician may take measurements by holding the probe electrode 1101 (to apply about 200 microamps to about 500 microamps) and ground electrode 1103 against the fur and/or skin at various measurement locations of the non-human animal. In other embodiments, the ground electrode 1103 and probe electrode 1101 may be configured to be attached to the body and/or limb of a non-human animal using a fastener, such as a strap.

During diagnostic sessions, the current sensor 122 takes multiple measurements of electrical current when the probe 1101 is positioned to deliver diagnostic electrical current at each measurement location for a predetermined diagnostic duration, such as about three to about five seconds. Put differently, the current sensor may take multiple measurements of electrical current at each point on the animal’s skin and/or fur when the probe electrode is placed. Each of these current measurements may be calibrated based on simultaneous pressure measurement(s) that is/are indicative of pressure or force exerted against the probe electrode or ground electrode. The calibration may assist to or attempt to eliminate the impact of varying pressure or force on the magnitude of the current readings and may be used to eliminate outlier measurements. Thus, for probe placement at each single measurement location, multiple calibrated, filtered current measurements may be taken. Then, measurements of electrical current may be taken, using the same method as described above for humans, and compared to the lookup table Table 1 also described above for diagnosis.

The foregoing is considered as illustrative only of the principles of the invention. Furthermore, because numerous modifications and changes will readily occur to those skilled in the art, it is not desired to limit the invention to the exact construction and operation shown and described. While the preferred embodiment has been described, the details may be changed without departing from the invention. 

What is claimed is:
 1. A method comprising: (a) taking, in a diagnostic session, a set of measurements of electric current that flow between a ground electrode and a probe electrode through a non-human animal body, the measurements being taken in such a way that: (i) different measurements in the set are taken while the probe electrode touches the non-human animal at different Prototype Measurement Locations, one location at a time, and (ii) each of the respective measurements in the set is taken while: (A) the ground electrode touches the non-human animal, and (B) the probe electrode touches a different area of the non-human animal at one of the Prototype Measurement Locations; (b) calculating, based on the set of measurements, a Prototype Electrical Current State for the diagnostic session; (c) employing a lookup table to identify a medical condition that the lookup table associates with the Prototype Electrical Current State, wherein the medical condition is a Prototype Medical Condition; and (d) outputting: (i) a diagnosis that the non-human animal has the medical condition, or (ii) a recommendation that the non-human animal undergo medical testing to evaluate whether the non-human animal has the medical condition.
 2. The method of claim 1, wherein the Prototype Medical Condition is a Class B Condition.
 3. The method of claim 1, wherein the Prototype Medical Condition is a Class M Condition.
 4. The method of claim 1, wherein the Prototype Medical Condition is a Class N Condition.
 5. The method of claim 1, wherein the Prototype Medical Condition is a Class P Condition.
 6. The method of claim 1, wherein the Prototype Medical Condition is a Class A Condition.
 7. The method of claim 1, wherein the Prototype Medical Condition is a Class C Condition.
 8. The method of claim 1, wherein the Prototype Medical Condition is a viral infection.
 9. The method of claim 1, wherein the Prototype Medical Condition is a bacterial infection.
 10. The method of claim 1, wherein the Prototype Medical Condition is a Class D Condition.
 11. A method comprising: (a) calculating a Prototype Electrical Current State for a diagnostic session, based on a set of measurements of electric current in a non-human animal; (b) employing a lookup table to identify a medical condition that the lookup table associates with the Prototype Electrical Current State, wherein the medical condition is a Prototype Medical Condition; and (c) outputting (i) a diagnosis that the non-human animal has the medical condition, or (ii) a recommendation that the non-human animal undergo medical testing to evaluate whether the non-human animal has the medical condition.
 12. The method of claim 11 wherein different measurements in the set were taken while: a probe electrode touched skin of the non-human animal at different Prototype Measurement Locations, one location at a time; and the electric current flowed between a ground electrode and the probe electrode through the non-human animal body.
 13. The method of claim 11, wherein each of the respective measurements in the set was taken while: the electric current flowed between a ground electrode and a probe electrode through the non-human animal’s body; the ground electrode touched skin of a non-human animal; and the probe electrode touched a different portion of skin of the non-human animal at a Prototype Measurement Location.
 14. The method of claim 11, wherein the Prototype Medical Condition is a Class B Condition, Class M Condition, Class N Condition or Class P Condition.
 15. A system comprising: (a) a current sensor that includes a ground electrode and a probe electrode; and (b) one or more computers, wherein (i) the current sensor is configured to take, during a diagnostic session, a set of measurements of electric current, in such a way that (A) the electric current being measured flows between the ground electrode and the probe electrode through a non-human animal body, (B) different measurements in the set are taken while the probe electrode touches skin of the non-human animal at different Prototype Measurement Locations, one location at a time, and (C) each of the respective measurements in the set is taken while (I) the ground electrode touches skin of the non-human animal, and (II) the probe electrode touches skin at one of the Prototype Measurement Locations of the non-human animal, and (ii) the one or more computers are programmed to: (A) calculate, based on the set of measurements, a Prototype Electrical Current State for the diagnostic session, (B) employ a lookup table to identify a medical condition that the lookup table associates with the Prototype Electrical Current State, wherein the medical condition is a Prototype Medical Condition, and (C) output (I) a diagnosis that the non-human animal has the medical condition, or (II) a recommendation that the non-human animal undergo medical testing to evaluate whether the non-human animal has the medical condition.
 16. The system of claim 15, wherein the ground electrode and the probe electrode are parts of a single rigid structure and are in a fixed position relative to each other.
 17. The system of claim 15, further comprising: one or more pressure sensors that are each configured to measure pressure exerted on the ground electrode or the probe electrode, wherein the ground electrode and the probe electrode are in fixed positions relative to each other, except for any movement due to displacement that occurs within the one or more pressure sensors.
 18. The system of claim 15, further comprising: an electronic display screen and an audio transducer, wherein the one or more computers are programmed to cause the screen and the audio transducer to together output an audiovisual presentation that provides information about whether the ground and probe electrodes are positioned correctly on the non-human animal.
 19. The system of claim 15, wherein the ground electrode and the probe electrode are parts of a single rigid structure and are in a fixed position relative to each other, and the rigid structure is configured to partially surround a smartphone or other mobile computing device.
 20. The system of claim 15, wherein the Prototype Medical Condition is a Class B Condition, Class M Condition, Class N Condition or Class P Condition. 